Electromagnetic flowmeter features
Using intelligent judgment, the measurement correction setting is not used, and the air traffic control alarm and electrode detection application are more convenient;
Advanced "rough error handling" technology, which can remove fluids such as slurry to measure sharp disturbances, reduce output runout, maintain high precision measurement and make output more stable;
With a fluid density setting, it can display mass flow;
Constant current excitation current range, 125mA, 250mA optional, can be used with different manufacturers, different types of electromagnetic flow sensors;
Control function with remote reset of the totalizer, with contact signal input for starting and stopping accumulation, suitable for total inspection and batch processing applications;
With self-test and self-diagnosis function;
Advanced non-volatile memory for higher circuit reliability and effective protection of setup and measurement parameters;
The meter can be equipped with an unpowered clock and memory for recording the power down time, power-on time and power-down time;
The meter has an optional hour recording function that can store flow and electrode resistance measurements for more than 30 days.
The new keyboard processing method avoids the keyboard operation affecting the measurement, and can enter and return the operation menu to make the parameter setting more convenient;
The total display uses 10-bit decimal 9999999999 full-value carry, which solves the practice of double-word full value 4294967285 (hexadecimal FFFF) carry-in habit;
With wireless transmission, the mesh network is organized by the wireless HART protocol.
Gas turbine flowmeter and screwing vortex flowmeter measurement difference
Keywords: gas turbine flowmeter, Tianxin gas turbine flowmeter, gas gas turbine flowmeter
First, the characteristics of gas turbine flowmeter and spiral vortex flowmeter
Gas turbine flowmeter and screwing vortex flowmeter are the main gas metering instruments in China. In foreign countries, almost no one uses gas to swirl into the vortex flow meter. At present, there are certain applications in the domestic vortex flowmeter. In terms of standard conditions. The international standard ISO9551 promulgated by the International Organization for Standardization and the international recommendations R6, R32 promulgated by the International Organization of Legal Metrology (OIML) have corresponding national standards and regulations in China and developed countries. The above documents guarantee the authority of gas turbine flowmeters in gas metering. Sex. Gas-introducing vortex flowmeters have not yet established relevant standards. It is difficult to guarantee the authority of the flow meter in gas metering. At present, in domestic flow meters for gas measurement, gas turbine flowmeters dominate, and the vortex flowmeters also have a certain market. The use of the two is similar, and users often have doubts about the choice of the two.
Liquid turbine flow meter classification and function description
Keywords: liquid turbine flowmeter, diesel liquid turbine flowmeter, pure water liquid turbine flowmeter
First, the liquid turbine flowmeter overview
The liquid turbine flowmeter is the main type of impeller flow (flow rate) meter, and the impeller type flowmeter also has an anemometer and a water meter. The TUF consists of a sensor and a conversion display. The sensor uses a multi-blade rotor to sense the average flow rate of the fluid, thereby deriving the flow or total amount. The rotational speed (or number of revolutions) of the rotor can be detected mechanically, magnetically, and photoelectrically and displayed and transmitted by the reading device.
Second, liquid turbine flowmeter classification
According to the classification of instrument functions, SDLWGY series turbine flowmeters can be divided into 2 categories.
1, turbine flow sensor / transmitter
2, intelligent integrated turbine flowmeter
Third, the liquid turbine flowmeter function description
1, turbine flow sensor / transmitter
This type of turbine flow product itself does not have a live display function, only the flow signal is transmitted far. The flow signal can be divided into pulse signal or current signal (4-20mA); the instrument is low in price, high in integration and small in size, and is especially suitable for use with computer control systems such as secondary display, PLC, DCS.