Because all kinds of ultrasonic flowmeters can be installed outside the pipe, non-contact flow measurement, the cost of the instrument is basically independent of the size of the pipe to be tested, while other types of flowmeters increase with the increase in caliber, so the cost is increased. The flowmeter is superior to the other functions of the same type of flowmeter. It is considered to be a good large-diameter flow measuring instrument. The Doppler ultrasonic flowmeter can measure the flow of two-phase medium, so it can be used for the measurement of dirty sewage such as sewers and sewage. In power plants, the use of portable ultrasonic flowmeters to measure large pipe diameters such as turbine water inflow and turbine circulating water is much more convenient than in the past. Ultrasonic flow juice can also be used for gas measurement. Pipe diameters range from 2cm to 5m, from a few meters wide open channels, culverts to 500m wide rivers.
In addition, the accuracy of the flow measurement of the ultrasonic measuring instrument is almost independent of the temperature, pressure, viscosity, density and other parameters of the measured fluid, and can be made into non-contact and portable measuring instruments, so it can solve the problem that other types of instruments are difficult to measure. Flow measurement problems for corrosive, non-conductive, radioactive, and flammable and explosive media. In addition, in view of the non-contact measurement characteristics, coupled with reasonable electronic circuits, one instrument can adapt to a variety of pipe diameter measurements and a variety of flow range measurements. The adaptability of ultrasonic flowmeters is also unmatched by other instruments. Ultrasonic flowmeters have some of the above advantages, so it has received more and more attention and has been developed into a series of products and generalization. It has been made into standard, high-temperature, explosion-proof and wet instruments of different channels to adapt to different media. Flow measurement for occasions and different pipeline conditions.
Second, gas turbine flowmeter measurement characteristics
1. Gas turbine flowmeter is one of the main types of flowmeters. Its working principle is: the gas entering the gas turbine flowmeter is accelerated by the fluid, and the accelerated gas acts on the blades of the turbine through the inlet passage. The impulse causes the turbine of the flow meter to rotate in the flow path, and within a certain flow range, the number of revolutions of the turbine is proportional to the volume of the gas flowing therethrough. By mechanical or electromagnetic induction, the number of revolutions f of the turbine can be obtained, so that the flow rate Q of the gas flowing through the flow meter can be obtained.
2. Gas turbine flowmeters are widely used in the measurement of gas flow or total gas volume, and are widely used in natural gas flow measurement at home and abroad. The theoretical research and practical experience of gas turbine flowmeters are very mature, and there are corresponding international standards. Countries also have corresponding national standards. China also adopts international standards. In Europe and the United States, gas turbine flow meters are the most important gas flow meters, and the Netherlands alone uses more than 2,600 gas turbine meters on natural gas pipelines.
Liquid turbine flow meter classification and function description
Keywords: liquid turbine flowmeter, diesel liquid turbine flowmeter, pure water liquid turbine flowmeter
First, the liquid turbine flowmeter overview
The liquid turbine flowmeter is the main type of impeller flow (flow rate) meter, and the impeller type flowmeter also has an anemometer and a water meter. The TUF consists of a sensor and a conversion display. The sensor uses a multi-blade rotor to sense the average flow rate of the fluid, thereby deriving the flow or total amount. The rotational speed (or number of revolutions) of the rotor can be detected mechanically, magnetically, and photoelectrically and displayed and transmitted by the reading device.
Second, liquid turbine flowmeter classification
According to the classification of instrument functions, SDLWGY series turbine flowmeters can be divided into 2 categories.
1, turbine flow sensor / transmitter
2, intelligent integrated turbine flowmeter
Third, the liquid turbine flowmeter function description
1, turbine flow sensor / transmitter
This type of turbine flow product itself does not have a live display function, only the flow signal is transmitted far. The flow signal can be divided into pulse signal or current signal (4-20mA); the instrument is low in price, high in integration and small in size, and is especially suitable for use with computer control systems such as secondary display, PLC, DCS.