The mediums that need to be measured in the site are different. Now, a brief introduction is made on the selection of flow meters for different media.
The selection of large water flow meter: suitable for electromagnetic flowmeter, ultrasonic flowmeter. Other applications such as vortex flowmeters, orifice flowmeters, etc. have faded out of this application, and these two types of flow meters are the most important in the industry. In particular, it should be mentioned that the application of Kaifeng Shengda's plug-in flowmeter and electromagnetic flowmeter are better.
Sewage, pulp and other turbid liquid meter selection: electromagnetic flowmeter can be used. However, when choosing an electromagnetic flowmeter, consider that the liquid does not contain more air or bubbles.
The meter with more bubbles of liquid selection: Doppler ultrasonic flowmeter can be selected, using this type of flowmeter to measure the fluid with bubbles, the effect is very good.
Flow meter selection with low conductivity such as pure water and demineralized water: Turbine flow meter is very suitable for measuring such fluids.
Flow meter selection for strong corrosive media such as acid and lye: 1. Electromagnetic flowmeter with acid and alkali lining. 2. clamp type ultrasonic flowmeter.
Selection of flowmeters for mediums with large concentration and large solid particle content such as mortar and electric powder slurry: electromagnetic flowmeter.
Flow meter selection for oil media such as oil and diesel: turbine flow meter and ultrasonic flow meter.
Gas flow meter selection: vortex flowmeter.
Advantages of turbine flow meters
The pressure loss is small, and the impeller has anti-corrosion function;
1.Adopt advanced ultra-low power single-chip microcomputer technology, the whole machine has strong functions, low power consumption and superior performance.
2.Intelligent flow display with nonlinear accuracy compensation. Correction formula accuracy is better than ±0.02%
3.The meter factor can be set online by button and can be displayed on the LCD screen. The LCD screen is intuitive and clear, and the reliability is strong.
4.EEPROM is used to accumulate the flow and the meter coefficient is powered down. The protection time is more than 10 years.
5.Using high-performance MCU central processing unit to complete data acquisition and processing display output, cumulative flow instantaneous flow
6.The screen displays a convenient man-machine interface for data transfer in standard 485 format.
7.Fully hard alloy (tungsten carbide) shielded cantilever beam structure bearing, integrating rotary bearing and pressure bearing, greatly improving bearing life and working in medium with a small amount of sediment and dirt.
8.It adopts 1Cr18Ni9Ti all stainless steel structure (the turbine adopts 2Cr13) and has good anti-corrosion performance.
9.Easy to maintain, self-rectifying structure, small and light, simple structure, which can be combined in a short time.
10.Internal cleaning is simple.
11.Strong anti-magnetic interference and vibration capability, reliable performance and long life
12.The lower limit flow rate is low and the measuring range is wide.
13.The on-site display type LCD screen is clear and intuitive, with low power consumption. The 3V lithium battery can be operated continuously for more than 5 years.
Corrosion resistant, suitable for acid and alkali solutions
Gas turbine flowmeter and screwing vortex flowmeter measurement difference
Keywords: gas turbine flowmeter, Tianxin gas turbine flowmeter, gas gas turbine flowmeter
First, the characteristics of gas turbine flowmeter and spiral vortex flowmeter
Gas turbine flowmeter and screwing vortex flowmeter are the main gas metering instruments in China. In foreign countries, almost no one uses gas to swirl into the vortex flow meter. At present, there are certain applications in the domestic vortex flowmeter. In terms of standard conditions. The international standard ISO9551 promulgated by the International Organization for Standardization and the international recommendations R6, R32 promulgated by the International Organization of Legal Metrology (OIML) have corresponding national standards and regulations in China and developed countries. The above documents guarantee the authority of gas turbine flowmeters in gas metering. Sex. Gas-introducing vortex flowmeters have not yet established relevant standards. It is difficult to guarantee the authority of the flow meter in gas metering. At present, in domestic flow meters for gas measurement, gas turbine flowmeters dominate, and the vortex flowmeters also have a certain market. The use of the two is similar, and users often have doubts about the choice of the two.