The mediums that need to be measured in the site are different. Now, a brief introduction is made on the selection of flow meters for different media.
The selection of large water flow meter: suitable for electromagnetic flowmeter, ultrasonic flowmeter. Other applications such as vortex flowmeters, orifice flowmeters, etc. have faded out of this application, and these two types of flow meters are the most important in the industry. In particular, it should be mentioned that the application of Kaifeng Shengda's plug-in flowmeter and electromagnetic flowmeter are better.
Sewage, pulp and other turbid liquid meter selection: electromagnetic flowmeter can be used. However, when choosing an electromagnetic flowmeter, consider that the liquid does not contain more air or bubbles.
The meter with more bubbles of liquid selection: Doppler ultrasonic flowmeter can be selected, using this type of flowmeter to measure the fluid with bubbles, the effect is very good.
Flow meter selection with low conductivity such as pure water and demineralized water: Turbine flow meter is very suitable for measuring such fluids.
Flow meter selection for strong corrosive media such as acid and lye: 1. Electromagnetic flowmeter with acid and alkali lining. 2. clamp type ultrasonic flowmeter.
Selection of flowmeters for mediums with large concentration and large solid particle content such as mortar and electric powder slurry: electromagnetic flowmeter.
Flow meter selection for oil media such as oil and diesel: turbine flow meter and ultrasonic flow meter.
Gas flow meter selection: vortex flowmeter.
Flowmeter product features:
Using intelligent judgment, the measurement correction setting is not used, and the air traffic control alarm and electrode detection application are more convenient;
Advanced "rough error handling" technology, which can remove fluids such as slurry to measure sharp disturbances, reduce output runout, maintain high precision measurement and make output more stable;
With a fluid density setting, it can display mass flow;
Constant current excitation current range, 125mA, 250mA optional, can be used with different manufacturers, different types of electromagnetic flow sensors;
Control function with remote reset of the totalizer, with contact signal input for starting and stopping accumulation, suitable for total inspection and batch processing applications;
With self-test and self-diagnosis function;
Advanced non-volatile memory for higher circuit reliability and effective protection of setup and measurement parameters;
The meter can be equipped with an unpowered clock and memory for recording the power down time, power-on time and power-down time;
The meter has an optional hour recording function that can store flow and electrode resistance measurements for more than 30 days.
The new keyboard processing method avoids the keyboard operation affecting the measurement, and can enter and return the operation menu to make the parameter setting more convenient;
The total display uses 10-bit decimal 9999999999 full-value carry, which solves the practice of double-word full value 4294967285 (hexadecimal FFFF) carry-in habit;
With wireless transmission, the mesh network is organized by the wireless HART protocol.
Third, the spiral vortex flowmeter measurement characteristics
The spiral vortex flowmeter is a flowmeter that began to appear in the 1970s. Its working principle is: the gas that enters the gas swirling vortex flowmeter is first forced by the spiral spinner to accelerate the rotation to form a vortex, the center of the vortex. For the vortex core. The accelerated vortex enters the enlarged section and then decelerates sharply. The pressure rises to produce a recirculation. Under the action of the recirculation, the vortex core makes a spiral precession around the axis of the flowmeter. The flow rate Q can be derived by measuring the vortex precession frequency f by sensing the sensitive component.
Fourth, gas turbine flowmeter and spiral vortex flowmeter measurement difference
The gas turbine flowmeter has a small pressure loss and can be suitable for gas metering in low pressure conveying applications. Gas-injected vortex flowmeters have a slightly higher pressure loss, and gas metering in low-pressure delivery situations sometimes has problems.
Gas turbine flowmeters require high media cleanliness and can be damaged if used improperly. The gas swirling vortex flowmeter gas swirling vortex flowmeter requires no gas turbine flowmeter and is not damaged.
Gas turbine flowmeters are highly accurate. It is the most accurate of all flow meters. The gas spiraling vortex flowmeter is not as accurate as the former.