The choice of electromagnetic flowmeter is mainly the correct choice of the transmitter, and the converter only needs to be matched with it.
Choice of caliber and range
The diameter of the transmitter is usually the same as that of the piping system. If the piping system is to be designed, the diameter can be selected according to the flow range and flow rate. For electromagnetic flowmeters, the flow rate is suitable for 2 - 4m / s. In special cases, such as solid particles in the liquid, in consideration of wear, the optional flow rate is ≤ 3m / s, for easy to manage the fluid. Available flow rate ≥ 2m / s. After the flow rate is determined, the transmitter diameter can be determined according to qv= D2.
The range of the transmitter can be selected according to two principles: one is that the full scale of the meter is greater than the expected maximum flow value; the other is that the normal flow is greater than 50% of the full scale of the meter to ensure a certain measurement accuracy.
Temperature and pressure selection
There are certain restrictions on the fluid pressure and temperature that the electromagnetic flowmeter can measure. When used, the pressure used must be lower than the working pressure specified by the flowmeter. At present, the working pressure specifications of domestically produced electromagnetic flowmeters are:
Less than 50mm caliber, working pressure is 1.6MPa; 900 mm caliber, working pressure is 1 MPa;More than 1000mm caliber, working pressure is 0.6MPa.
If there are special requirements on the pressure resistance of the transmitter, you can negotiate with the manufacturer.
The operating temperature of the electromagnetic flowmeter depends on the lining material used, which is generally 5 - 70 ° C. Such as special treatment, can exceed the above range, such as the wear-resistant corrosion-resistant electromagnetic flowmeter produced by Tianjin Automation Instrument No.3. The transmitter allows the measured medium temperature to be -40 to 130 °C.
Selection of lining material and electrode tree material
The lining material and electrode material of the transmitter must be correctly selected according to the physical and chemical properties of the medium. Otherwise, the instrument will be damaged quickly due to the corrosion of the lining and the electrode, and the corrosive medium will easily cause an accident if it leaks. Therefore, the electrode and the lining material must be carefully selected according to the specific measurement medium in the production process.
The main technical parameters of ultrasonic external clamp flowmeter:
Measurement accuracy: better than 1%
Repeatability: better than 0.2%
Measurement period: 500ms (2 times per second, 128 sets of data collected per cycle)
Working power: 220VAC/8~36VDC
Maximum flow rate: 64m / s (flow rate resolution 0.001m / s)
Display: 2 × 10 Chinese characters backlit LCD can display instantaneous flow and positive, negative, net cumulative flow, flow rate, etc.
Operation: 4×4 touch keyboard (F4 main unit magnetic 4 button) operation
Signal input: 3 channels of 4-20mA analog input, accuracy 0.1%, can input pressure, liquid level, temperature and other signals
2-way three-wire PT100 platinum resistor
Signal output: 1 isolated RS485 output
1 way 4-20mA or 0-20mA output
1 channel isolated OCT (programmable between pulse width 6~1000ms, default 200ms)
1 relay output (pulse width 200ms)
Data storage: Optional built-in data storage (SD card) can store time, instantaneous flow, accumulated flow, signal status, etc., through special software, can import data into computer for statistical and management
Communication protocol: MODBUS protocol, M-BUS protocol, FUJI extension protocol, and compatible with communication protocols of similar products of other domestic manufacturers
Other functions: automatic memory before 512 days, the first 128 months, the first 10 years of positive / negative / net cumulative flow
Automatically remember the first 30 times of power-on and power-off time and flow rate, and can realize automatic or manual addition of traffic, which can be read by MODBUS protocol.
Programmable batch (quantitative) controller, fault self-diagnosis function
Software upgrade via code file transmitted via E-mail
Protection level: sensor IP68, F4 host IP68, remaining host IP65
Explosion-proof grade: EXdIIBT4 (JN-100F2 type)
Gas turbine flowmeter and screwing vortex flowmeter measurement difference
Keywords: gas turbine flowmeter, Tianxin gas turbine flowmeter, gas gas turbine flowmeter
The gas turbine flowmeter has good repeatability and is the preferred flow meter for trade settlement. The gas swirling vortex flowmeter is generally repeatable.
The gas turbine flowmeter can obtain very high frequency signals with strong signal resolution. The frequency of the gas swirling vortex flowmeter is very low, and the smaller the diameter, the lower the signal resolution. The measurement error is to be measured.
The gas turbine flowmeter has a wide range, generally 40:1~20:1, and has a wide measuring range, which can be suitable for occasions with large flow changes. When natural gas is delivered, the flow changes are generally large and, therefore, are particularly suitable for natural gas measurements. The latter range is medium, generally 15:1 to 10:1, and the measurement range is slightly narrower. When measuring gas, it is more suitable for occasions where the flow rate does not change much.
In terms of seismic performance, gas turbine flowmeters have better resistance to pipeline vibration. The gas swirling vortex flowmeter is susceptible to pipeline vibration and should be taken care of when using it.