Choice of caliber
The choice of the diameter of the electromagnetic flowmeter is not necessarily the same as the diameter of the pipe, and should be determined by the flow rate and flow rate. However, as the chemical raw materials and intermediate liquid of Salt Lake, the viscosity is large and the flow rate is low (generally 2.0 to 4.0 m/s). The electromagnetic flowmeter is used on such a pipe, and the diameter of the sensor is the same as the diameter of the pipe. If the flow rate is lower than 1.0m/s, an electromagnetic flowmeter of appropriate flow rate can be selected to ensure measurement accuracy and relatively reduce investment.
Selection of flow rate and range
Basically, they are anti-corrosion plastic pipes, and considering the lining of the flow meter, the flow rate is generally controlled at 2m/s. For some materials that are easy to crystallize (such as sodium carbonate solution, compounding liquid: crystallized mainly boron and magnesium double salt), the flow rate is increased to 3.0 to 4.0 m/s by taking measures. The flow rate will increase the flow noise, and the vibration of the pipeline will affect the measurement accuracy. Under the condition that the electromagnetic flowmeter is installed, the shock absorber should be installed before and after. The full scale of the meter is greater than the expected maximum flow value, which is typically 1.2 times the estimated maximum flow. The normal measurement flow is greater than 50% of the full scale of the meter to ensure a certain measurement accuracy.
Turbine flowmeter composition
Turbine flowmeters are speed flowmeters, also known as impeller flowmeters. The impeller type flowmeter utilizes the relationship between the rotational angular velocity of the impeller placed in the fluid and the fluid flow velocity. By measuring the rotational speed of the impeller to reflect the volumetric flow rate of the fluid passing through the pipeline, it is a relatively mature high-accuracy instrument in the flowmeter. One. Turbine flowmeters generally consist of the following five typical parts:
The material of the body is generally steel or cast iron, and the ends are flanged. The small diameter gauge also has a threaded interface.
2. Measured component
The turbine has precision machined blades that together with a set of reduction gears and bearings form the measuring assembly. The two high-precision stainless steel permanent self-lubricating bearings that support the turbine ensure a long service life of the assembly. Turbine flowmeters can also be lubricated with an external lubricant pump, but be careful not to overdo it.
The following important information is available on the counter panel:
(1) Maximum working temperature and pressure:
(2) Metering and minimum and maximum flow levels
(3) Product model and number
(4) Explosion-proof grades and signs
(5) The equivalent of the fluid corresponding to the low frequency or high frequency pulse and the wiring method.
The rectifier is used to keep the fluid flowing through the turbine flow meter in a regular state, thereby eliminating the disturbance and having an unaffected effect on the metering. High metering accuracy.
5. Magnetic coupling transmission device
The device separates the counter portion in the atmospheric environment from the gas being measured and transmits the rotation of the measuring assembly to the counter.
Gas turbine flowmeter and screwing vortex flowmeter measurement difference
Keywords: gas turbine flowmeter, Tianxin gas turbine flowmeter, gas gas turbine flowmeter
The gas turbine flowmeter has good repeatability and is the preferred flow meter for trade settlement. The gas swirling vortex flowmeter is generally repeatable.
The gas turbine flowmeter can obtain very high frequency signals with strong signal resolution. The frequency of the gas swirling vortex flowmeter is very low, and the smaller the diameter, the lower the signal resolution. The measurement error is to be measured.
The gas turbine flowmeter has a wide range, generally 40:1~20:1, and has a wide measuring range, which can be suitable for occasions with large flow changes. When natural gas is delivered, the flow changes are generally large and, therefore, are particularly suitable for natural gas measurements. The latter range is medium, generally 15:1 to 10:1, and the measurement range is slightly narrower. When measuring gas, it is more suitable for occasions where the flow rate does not change much.
In terms of seismic performance, gas turbine flowmeters have better resistance to pipeline vibration. The gas swirling vortex flowmeter is susceptible to pipeline vibration and should be taken care of when using it.