Choice of caliber
The choice of the diameter of the electromagnetic flowmeter is not necessarily the same as the diameter of the pipe, and should be determined by the flow rate and flow rate. However, as the chemical raw materials and intermediate liquid of Salt Lake, the viscosity is large and the flow rate is low (generally 2.0 to 4.0 m/s). The electromagnetic flowmeter is used on such a pipe, and the diameter of the sensor is the same as the diameter of the pipe. If the flow rate is lower than 1.0m/s, an electromagnetic flowmeter of appropriate flow rate can be selected to ensure measurement accuracy and relatively reduce investment.
Selection of flow rate and range
Basically, they are anti-corrosion plastic pipes, and considering the lining of the flow meter, the flow rate is generally controlled at 2m/s. For some materials that are easy to crystallize (such as sodium carbonate solution, compounding liquid: crystallized mainly boron and magnesium double salt), the flow rate is increased to 3.0 to 4.0 m/s by taking measures. The flow rate will increase the flow noise, and the vibration of the pipeline will affect the measurement accuracy. Under the condition that the electromagnetic flowmeter is installed, the shock absorber should be installed before and after. The full scale of the meter is greater than the expected maximum flow value, which is typically 1.2 times the estimated maximum flow. The normal measurement flow is greater than 50% of the full scale of the meter to ensure a certain measurement accuracy.
Working principle and sensor parameters of ultrasonic external clamp flow meter
External clip type ultrasonic flow meter external clip type ultrasonic flow meter series measuring full flow of liquid flow meter, optional explosion-proof and intrinsically safe type, explosion-proof grade Exd[ib]IIBT6, and optional intrinsically safe remote control, explosion-proof grade ExibIIBT6, measurable medium: drinking water, river water, sea water, ground water cooling water, high temperature water, sewage, lubricating oil, diesel, fuel chemical liquid, other homogeneous fluid.
External clip sensor parameters:
The external clip type sensor is an installation method in which the sensor is directly bundled on the outer surface of the pipe to be measured to realize flow measurement, and has a diameter independent of the pipe diameter.
Simple installation, no need to stop production, no pressure loss and so on.
Executive Standard: JB/T 9248-1999
Nominal diameter: 15, 20, 25, 32, 40, 50, 65, 80, 100, 125, 150, 200, 250, 300, 350, 400, 450, 500, 600, 700, 800, 900, 1000, 1200, 1400, 1600, 1800, 2000, 2200, 2400, 2600, 2800, 3000
Maximum flow rate: 15m/s
Accuracy: DNl5~DN600, indication: ±0.3% (flow rate ≥1m/s); ±3mm/s (flow rate <1m/s)
DN700-DN3000, ±0.5% of the indicated value (flow rate ≥0.8m/S); ±4mm/s (flow rate <0.8m/S)
Fluid conductivity ≥5uS/cm
DNl5～DN: 1504.0MPa, DNl5～DN600: 1.6MPa, DN200～DN1000: 1.0MPa, DN700～DN3000: 0.6MPa, Special order: 6.3, 10MPa
Sensor: -25 °C - ten 60 °C
Converter and integrated type: -10 ° C - ten 60 °C
Lining material: PTFE, neoprene, polyurethane, polytetrafluoroethylene (F46), screened PFA
Maximum fluid temperature
- Body type 70 °C
Separate type: Polychloroprene lining 80 ° C; 120 °C (specify when ordering)
Polyurethane lining 80 °C
Polytetrafluoroethylene propylene (F46) 100 °C; 150 °C (specify when ordering)
Signal electrode and grounding electrode material: stainless steel 0Crl8Nil2M02Ti, Hastelloy C, Hastelloy B, titanium, tantalum, platinum/rhodium alloy, stainless steel coated tungsten carbide
Electrode scraper mechanism: DN300-DN3000
Connecting flange material: carbon steel
Grounding flange material: stainless steel 1Crl8Ni9Ti
DN65—DNl50: Stainless steel 1Crl8Ni9Ti
DN200～DNl600: Carbon steel ten stainless steel 1Crl8Ni9Ti
DNl5～DN3000 separate rubber or polyurethane lining sensor: IP65 or IP68
Other sensors, body flow meters and split converters: IP65
Spacing (separate type): The converter distance sensor generally does not exceed 100m