Selection of electrodes and lining
Standard electrodes are used for media that do not produce non-staining electrodes such as crystals, agglomerates, and scales. Otherwise, a doctor blade electrode is used. In addition to the pure alkaline solution, the dosing solution (crystallization of boron and magnesium double salt), the washing liquid and other electromagnetic flowmeters that use the scraper electrode, the other uses the electromagnetic flowmeter of the standard electrode. Since the process pipes in the workshop are basically made of plastic insulation material, if the electromagnetic flowmeter with two electrodes is selected, the grounding ring must be configured to ensure the measurement accuracy. However, the common grounding ring is easy to corrode, and the corrosion-resistant titanium or 钽 grounding ring is high in cost, so the three-electrode electromagnetic flowmeter is selected to ensure measurement accuracy, reduce cost, and convenient installation and maintenance.
Flowmeter type selection
The types of electromagnetic flowmeters are divided into: integral type and split type. In Qinghai, the coldest temperature in winter is up to -30 °C. Many metered electromagnetic flowmeters are outdoors, and the environmental humidity of magnesium-lithium separation, pressure filtration and washing processes in lithium and boron systems is large. Exquisite, evaporation, conversion and other processes up to 90 °C. Therefore, the humidity of the high temperature, low temperature, high temperature fluid, vibration source and other working conditions select a split flow meter to effectively prevent the impact on the measuring components, to ensure measurement accuracy and the life of the electromagnetic flowmeter. However, it should be noted that the signal cable between the electromagnetic flowmeter sensor and the transmitter should not be too long, and the manufacturer-specific cable should be used, otherwise the measurement accuracy will be affected. The rest of the working conditions are more ideal.
Turbine flowmeter composition
Turbine flowmeters are speed flowmeters, also known as impeller flowmeters. The impeller type flowmeter utilizes the relationship between the rotational angular velocity of the impeller placed in the fluid and the fluid flow velocity. By measuring the rotational speed of the impeller to reflect the volumetric flow rate of the fluid passing through the pipeline, it is a relatively mature high-accuracy instrument in the flowmeter. One. Turbine flowmeters generally consist of the following five typical parts:
The material of the body is generally steel or cast iron, and the ends are flanged. The small diameter gauge also has a threaded interface.
2. Measured component
The turbine has precision machined blades that together with a set of reduction gears and bearings form the measuring assembly. The two high-precision stainless steel permanent self-lubricating bearings that support the turbine ensure a long service life of the assembly. Turbine flowmeters can also be lubricated with an external lubricant pump, but be careful not to overdo it.
The following important information is available on the counter panel:
(1) Maximum working temperature and pressure:
(2) Metering and minimum and maximum flow levels
(3) Product model and number
(4) Explosion-proof grades and signs
(5) The equivalent of the fluid corresponding to the low frequency or high frequency pulse and the wiring method.
The rectifier is used to keep the fluid flowing through the turbine flow meter in a regular state, thereby eliminating the disturbance and having an unaffected effect on the metering. High metering accuracy.
5. Magnetic coupling transmission device
The device separates the counter portion in the atmospheric environment from the gas being measured and transmits the rotation of the measuring assembly to the counter.
Liquid turbine flow meter classification and function description
Keywords: liquid turbine flowmeter, diesel liquid turbine flowmeter, pure water liquid turbine flowmeter
First, the liquid turbine flowmeter overview
The liquid turbine flowmeter is the main type of impeller flow (flow rate) meter, and the impeller type flowmeter also has an anemometer and a water meter. The TUF consists of a sensor and a conversion display. The sensor uses a multi-blade rotor to sense the average flow rate of the fluid, thereby deriving the flow or total amount. The rotational speed (or number of revolutions) of the rotor can be detected mechanically, magnetically, and photoelectrically and displayed and transmitted by the reading device.
Second, liquid turbine flowmeter classification
According to the classification of instrument functions, SDLWGY series turbine flowmeters can be divided into 2 categories.
1, turbine flow sensor / transmitter
2, intelligent integrated turbine flowmeter
Third, the liquid turbine flowmeter function description
1, turbine flow sensor / transmitter
This type of turbine flow product itself does not have a live display function, only the flow signal is transmitted far. The flow signal can be divided into pulse signal or current signal (4-20mA); the instrument is low in price, high in integration and small in size, and is especially suitable for use with computer control systems such as secondary display, PLC, DCS.