Pressure and temperature selection
The actual working pressure of the fluid in the pipeline should be less than the rated working pressure of the flowmeter. Pay attention to whether there is negative pressure in the pipeline, such as the evaporation process, and the negative pressure is up to -60KPa. Such working conditions should be communicated with the manufacturer's technicians when ordering. Strict requirements are imposed on the lining material. The actual temperature should meet the temperature requirements specified by the flowmeter.
Power supply selection
The choice of the electromagnetic flowmeter power supply should be selected according to the flowmeter itself and the environment of the site. If the site allows, the power supply is convenient, safe and easy to maintain. As a chemical company, more than 90% of the raw materials and process links are in the liquid phase. During operation, the internal humidity of the workshop is large, and no flammable gas is generated during the whole operation. It is safer to use DC24V for power supply. Outdoor dry environment, choose AC220V power supply, convenient lead wire, good maintenance and low cost.
Some explosion-proof occasions, such as flowmeters used in gas boilers, are ideal for battery-powered electromagnetic flowmeters.
Working principle and sensor parameters of ultrasonic external clamp flow meter
External clip type ultrasonic flow meter external clip type ultrasonic flow meter series measuring full flow of liquid flow meter, optional explosion-proof and intrinsically safe type, explosion-proof grade Exd[ib]IIBT6, and optional intrinsically safe remote control, explosion-proof grade ExibIIBT6, measurable medium: drinking water, river water, sea water, ground water cooling water, high temperature water, sewage, lubricating oil, diesel, fuel chemical liquid, other homogeneous fluid.
External clip sensor parameters:
The external clip type sensor is an installation method in which the sensor is directly bundled on the outer surface of the pipe to be measured to realize flow measurement, and has a diameter independent of the pipe diameter.
Simple installation, no need to stop production, no pressure loss and so on.
Second, gas turbine flowmeter measurement characteristics
1. Gas turbine flowmeter is one of the main types of flowmeters. Its working principle is: the gas entering the gas turbine flowmeter is accelerated by the fluid, and the accelerated gas acts on the blades of the turbine through the inlet passage. The impulse causes the turbine of the flow meter to rotate in the flow path, and within a certain flow range, the number of revolutions of the turbine is proportional to the volume of the gas flowing therethrough. By mechanical or electromagnetic induction, the number of revolutions f of the turbine can be obtained, so that the flow rate Q of the gas flowing through the flow meter can be obtained.
2. Gas turbine flowmeters are widely used in the measurement of gas flow or total gas volume, and are widely used in natural gas flow measurement at home and abroad. The theoretical research and practical experience of gas turbine flowmeters are very mature, and there are corresponding international standards. Countries also have corresponding national standards. China also adopts international standards. In Europe and the United States, gas turbine flow meters are the most important gas flow meters, and the Netherlands alone uses more than 2,600 gas turbine meters on natural gas pipelines.