Choice of protection level
The protection level of the electromagnetic flowmeter should be selected according to the actual situation. If the sensor is required to be installed below the ground and is often flooded, IP68 should be selected. If the sensor is installed above the ground, IP65 and IP67 should be selected. In any case, the display area of ??the electromagnetic flowmeter cannot be used with water or enter the water, which may cause damage to the electromagnetic flowmeter. As a chemical company, the workshop produces high humidity, but IP65 (IP65 is a water-proof type. The electromagnetic flowmeter sensor housing allows the faucet to spray water from the sensor in any direction of the sensor of the electromagnetic flowmeter. The pressure of the water spray is 30Kpa, the water output It is 12.5L/S and the distance is 3 meters.) The electromagnetic flowmeter of the protection grade fully meets the requirements of on-site measurement.
Choice of connection method
Electromagnetic flowmeters have threaded connections, flanged connections, clamps, etc. In the chemical industry, flange-connected electromagnetic flowmeters are generally used. The selection must be consistent with the process-flange engineering pressure and standards.
Conclusion, with the maturity of the domestic instrument industry, chemical companies such as lithium carbonate, boric acid and potash will also enter the market in line with the domestic market. As an important industrial flow measuring instrument, electromagnetic flowmeter should play its role in DCS control. The right selection is especially important to provide accurate measurement data, reduce labor costs and increase production efficiency.
Working principle and sensor parameters of ultrasonic external clamp flow meter
External clip type ultrasonic flow meter external clip type ultrasonic flow meter series measuring full flow of liquid flow meter, optional explosion-proof and intrinsically safe type, explosion-proof grade Exd[ib]IIBT6, and optional intrinsically safe remote control, explosion-proof grade ExibIIBT6, measurable medium: drinking water, river water, sea water, ground water cooling water, high temperature water, sewage, lubricating oil, diesel, fuel chemical liquid, other homogeneous fluid.
External clip sensor parameters:
The external clip type sensor is an installation method in which the sensor is directly bundled on the outer surface of the pipe to be measured to realize flow measurement, and has a diameter independent of the pipe diameter.
Simple installation, no need to stop production, no pressure loss and so on.
Third, the spiral vortex flowmeter measurement characteristics
The spiral vortex flowmeter is a flowmeter that began to appear in the 1970s. Its working principle is: the gas that enters the gas swirling vortex flowmeter is first forced by the spiral spinner to accelerate the rotation to form a vortex, the center of the vortex. For the vortex core. The accelerated vortex enters the enlarged section and then decelerates sharply. The pressure rises to produce a recirculation. Under the action of the recirculation, the vortex core makes a spiral precession around the axis of the flowmeter. The flow rate Q can be derived by measuring the vortex precession frequency f by sensing the sensitive component.
Fourth, gas turbine flowmeter and spiral vortex flowmeter measurement difference
The gas turbine flowmeter has a small pressure loss and can be suitable for gas metering in low pressure conveying applications. Gas-injected vortex flowmeters have a slightly higher pressure loss, and gas metering in low-pressure delivery situations sometimes has problems.
Gas turbine flowmeters require high media cleanliness and can be damaged if used improperly. The gas swirling vortex flowmeter gas swirling vortex flowmeter requires no gas turbine flowmeter and is not damaged.
Gas turbine flowmeters are highly accurate. It is the most accurate of all flow meters. The gas spiraling vortex flowmeter is not as accurate as the former.