Choice of protection level
The protection level of the electromagnetic flowmeter should be selected according to the actual situation. If the sensor is required to be installed below the ground and is often flooded, IP68 should be selected. If the sensor is installed above the ground, IP65 and IP67 should be selected. In any case, the display area of ??the electromagnetic flowmeter cannot be used with water or enter the water, which may cause damage to the electromagnetic flowmeter. As a chemical company, the workshop produces high humidity, but IP65 (IP65 is a water-proof type. The electromagnetic flowmeter sensor housing allows the faucet to spray water from the sensor in any direction of the sensor of the electromagnetic flowmeter. The pressure of the water spray is 30Kpa, the water output It is 12.5L/S and the distance is 3 meters.) The electromagnetic flowmeter of the protection grade fully meets the requirements of on-site measurement.
Choice of connection method
Electromagnetic flowmeters have threaded connections, flanged connections, clamps, etc. In the chemical industry, flange-connected electromagnetic flowmeters are generally used. The selection must be consistent with the process-flange engineering pressure and standards.
Conclusion, with the maturity of the domestic instrument industry, chemical companies such as lithium carbonate, boric acid and potash will also enter the market in line with the domestic market. As an important industrial flow measuring instrument, electromagnetic flowmeter should play its role in DCS control. The right selection is especially important to provide accurate measurement data, reduce labor costs and increase production efficiency.
Ultrasonic flowmeter performance characteristics
Measuring liquid: single fluid such as water, seawater, alkali, etc., turbidity less than 10000ppm, particle size less than 1mm
Pipe material: Ultrasonic penetrating full pipe such as steel, cast iron, PVC pipe, dense texture, no serious corrosion and scaling, allowing thin layer of dense lining
Pipe lining material: rubber, epoxy asphalt, fiberglass, mortar, etc., or unlined
Pipe diameter range: DN15mm~6000mm, plug-in sensor for pipe diameter DN25mm~6000mm
Flow rate range: -32m/s~0~+32m/s, bidirectional flow
Accuracy: ±1.0% of displayed value (under standard conditions). Configure plug-in sensor accuracy up to 0.5%
Signal output: 4-20mA or 0-20mA RS-485
Sensor: It is divided into external clip type and plug-in type. The standard configuration is clip-on. The maximum operating pressure of the plug-in sensor is 3 MPa.
Straight pipe length: upstream >10D, downstream >5D from pump outlet >30D (D: pipe diameter)
Display mode: 2*20 character LCD display with LED backlight
Protection level: IP65
Explosion-proof grade: ExdIIBT4
Fixed ultrasonic flowmeter power supply: AC110V, AC220V, DC8~36V or AC7~30V
Ambient temperature / humidity: Host -20 ° C ~ +80 ° C; sensor -20 ° C ~ +80 ° C or -20 ° C ~ +160 ° C optional, the sensor can be immersed in water, water depth less than 2m.
Gas turbine flowmeter and screwing vortex flowmeter measurement difference
Keywords: gas turbine flowmeter, Tianxin gas turbine flowmeter, gas gas turbine flowmeter
First, the characteristics of gas turbine flowmeter and spiral vortex flowmeter
Gas turbine flowmeter and screwing vortex flowmeter are the main gas metering instruments in China. In foreign countries, almost no one uses gas to swirl into the vortex flow meter. At present, there are certain applications in the domestic vortex flowmeter. In terms of standard conditions. The international standard ISO9551 promulgated by the International Organization for Standardization and the international recommendations R6, R32 promulgated by the International Organization of Legal Metrology (OIML) have corresponding national standards and regulations in China and developed countries. The above documents guarantee the authority of gas turbine flowmeters in gas metering. Sex. Gas-introducing vortex flowmeters have not yet established relevant standards. It is difficult to guarantee the authority of the flow meter in gas metering. At present, in domestic flow meters for gas measurement, gas turbine flowmeters dominate, and the vortex flowmeters also have a certain market. The use of the two is similar, and users often have doubts about the choice of the two.