Electromagnetic flowmeter features
Frequency programmable low frequency rectangular wave excitation improves stability of flow measurement and low power loss.
The new 16-bit ultra-low power microprocessor with FLASH memory has high integration, fast calculation speed and high calculation accuracy.
Full digital processing, strong anti-interference ability and reliable measurement.
Ultra-low EMI switching power supply, suitable for power supply voltage variation range, high efficiency, small temperature rise; good EMC performance.
Chinese and English menu operation, easy to use, easy to operate, easy to learn and understand.
High definition backlight wide temperature LCD display.
It can perform bidirectional flow measurement and bidirectional total accumulation; it has automatic range switching function, which can effectively improve the measurement accuracy of analog current and frequency output, especially suitable for occasions where the diurnal flow range changes greatly and needs to send control signals; flow measurement range Up to 1500:1.
There are three totalizers inside, which record and display the forward cumulative amount, the reverse cumulative amount and the accumulated difference integrated amount, which are convenient for fluid metering and custody transfer.
Provide isolated or non-isolated RS485/RS232C digital communication interface, and support fieldbus communication modes such as MODBUS, PROFIBUS-DP and HART.
Constant current source fluid resistance measurement can accurately measure the internal resistance of the electrode signal in the case of long-line transmission. It can be used not only to determine whether the fluid in the sensor is empty or not, but also to identify abnormal phenomena such as contamination and coverage of the electrode, and to provide cleaning for the user. Fault processing information such as electrodes.
Second, the compressed air vortex flowmeter installation steps
1. The special flanges to be equipped are welded to the upstream and downstream straight pipe sections respectively, so that the inner diameters of the special flange and the straight pipe section are strictly vertical and concentric.
2. Install the sensor clip on the upper and lower straight pipe sections welded with special flanges and fasten them with bolts. The upstream and downstream straight sections should be kept concentric with the sensor.
3. It should be noted that the flow direction of the sensor should be consistent with the flow direction of the fluid in the pipeline.
Third, special attention should be paid when installing compressed air vortex flowmeter
1. When welding special flanges and straight pipe sections, the sensor should be removed and the sensor welding flange must not be carried.
2. Before the sensor is installed, the sealing ring must be placed in the groove of the flange.
3. The pressure point and temperature measurement point should be at the 3DN~5DN and 6N~8DN downstream of the sensor.
4. When the high temperature pipeline is insulated, do not wrap the sensor to avoid damage.
5. The cable connecting the vortex flowmeter should be as far away as possible from the interference of strong electromagnetic fields. It is absolutely not allowed to lay with high voltage cables. The shielded cable should be as short as possible, the maximum length should not exceed 500 meters, and should not be coiled to reduce the distributed inductance.
Third, gas turbine flowmeter daily inspection
1. Look at the display: Observe the total amount, instantaneous, temperature, pressure, and battery power. If the abnormality is reported in time.
2. Listen to the running sound of the flowmeter. (Method) The flowmeter should have no noise when it is running normally. If there is a large abnormal noise, it can be judged as damage to the movement of the movement (for example). It needs to be reported and processed in time.
3, data comparison, the output of the first 3 to 5 days before the statistics, if there is no major change, the flowmeter runs normally, such as the successive decrement phenomenon, the gas turbine flowmeter impeller (shaft, shaft core), filtration There is a problem with the device. Need to be reported in time for processing
4, gas turbine flowmeter filling lubricant
(1) Instrument refueling is an important part of the daily maintenance work of the instrument. Lubricating oil is mainly for instrument bearing; lubricating oil can protect the instrument bearing, reduce friction, reduce wear and prolong the operation cycle of the instrument;
(2) Regularly adding lubricating oil can effectively reduce the maintenance rate of the instrument and reduce the maintenance cost;
(3) Ensure the measurement accuracy of the flowmeter. "Measuring is counting money." why would you said this? The management of production and operation accounting, energy saving, cost accounting, etc. is an important production link of enterprise management. If the measurement is not in place, it will affect the economic efficiency and reputation of the enterprise. Therefore, the daily maintenance of the instrument is very important.