Plug-in ultrasonic flowmeter features
Programmable frequency low frequency rectangular wave excitation improves stability of flow measurement and low power loss;
Using 16-bit embedded microprocessor, the operation speed is fast and the precision is high;
Full digital processing, strong anti-interference ability, reliable measurement, high precision, flow measurement range up to 100:1;
Ultra-low EMI switching power supply, suitable for wide range of power supply voltage and good anti-EMC performance;
Full Chinese character menu operation, easy to use, easy to operate, easy to learn and understand;
High definition backlit LCD display;
It has two-way flow measurement and two-way total accumulation function, and the current and frequency have bidirectional output function;
There are three totalizers in the internal to display the forward cumulative amount, the reverse cumulative amount and the difference integrated amount;
With RS485 or RS232 digital communication signal output
It has a conductivity measurement function to determine whether the sensor is empty or not;
The constant current excitation current range is large, and can be used with different companies and different types of electromagnetic flow sensors;
With self-test and self-diagnosis function;
High reliability with SMD devices and surface mount (SMT) technology;
The internal design of the meter has a power-down clock that can record 16 power-down times.
Steam vortex flowmeter measurement requirements
Keywords: steam vortex flowmeter, saturated steam vortex flowmeter, superheated steam vortex flowmeter
What should I pay attention to when measuring vortex flowmeter? What is the best way to measure steam installation by vortex flowmeter? Can vortex flowmeters be used for steam metering? The vortex flowmeters are evaluated for their measurement performance under single-phase flow conditions. There is currently no single-phase flowmeter for measuring the system variation of two-phase flow. However, the two-phase flow exists objectively, and its influence on flow measurement is unavoidable. Despite the difficulties, people are still trying to study the mechanism of its influence on flow measurement, and take corresponding countermeasures to improve the accuracy of flow measurement.
First, steam flow measurement difficulties
During long-distance transportation, dry steam will partially condense due to heat loss, resulting in reduced steam dryness and become wet steam. The gas-liquid two-phase flow structure in the horizontal pipe is related to the gas-liquid two-phase volume ratio and the flow velocity. In the steam pipe, since the volume ratio of the condensed water in the wet steam is small, the drain pipe drawn from the bottom of the horizontal pipe is made. , can receive a good hydrophobic effect. When the flow rate is particularly high, it will also behave as a circular flow, that is, there is a liquid film on the pipe wall, and the central part of the pipe is a gas core with droplets. Due to the influence of gravity during horizontal flow, the lower liquid film is higher than the upper pipe. The thickness of the wall, in the vertical ascending pipeline, the basic structure of the gas-liquid two-phase flow has a fine bubble flow structure, a bullet-like flow structure, a block flow structure, a ring-shaped flow structure with fibers, and an annular flow structure.
Second, the compressed air vortex flowmeter installation steps
1. The special flanges to be equipped are welded to the upstream and downstream straight pipe sections respectively, so that the inner diameters of the special flange and the straight pipe section are strictly vertical and concentric.
2. Install the sensor clip on the upper and lower straight pipe sections welded with special flanges and fasten them with bolts. The upstream and downstream straight sections should be kept concentric with the sensor.
3. It should be noted that the flow direction of the sensor should be consistent with the flow direction of the fluid in the pipeline.
Third, special attention should be paid when installing compressed air vortex flowmeter
1. When welding special flanges and straight pipe sections, the sensor should be removed and the sensor welding flange must not be carried.
2. Before the sensor is installed, the sealing ring must be placed in the groove of the flange.
3. The pressure point and temperature measurement point should be at the 3DN~5DN and 6N~8DN downstream of the sensor.
4. When the high temperature pipeline is insulated, do not wrap the sensor to avoid damage.
5. The cable connecting the vortex flowmeter should be as far away as possible from the interference of strong electromagnetic fields. It is absolutely not allowed to lay with high voltage cables. The shielded cable should be as short as possible, the maximum length should not exceed 500 meters, and should not be coiled to reduce the distributed inductance.