The structure of the electromagnetic flowmeter is mainly composed of a magnetic circuit system, a measuring conduit, an electrode, a casing, a lining, and a converter.
Magnetic circuit system: its role is to produce a uniform DC or AC magnetic field. The DC magnetic circuit is realized by a permanent magnet, which has the advantages of simple structure and less interference by the alternating magnetic field, but it is easy to polarize the electrolyte liquid in the measuring duct, so that the positive electrode is surrounded by negative ions, and the negative electrode is positive ion Surrounding, that is, the polarization phenomenon of the electrode, and causing an increase in internal resistance between the two electrodes, thus seriously affecting the normal operation of the meter. When the diameter of the pipe is large, the permanent magnets are correspondingly large, bulky and uneconomical, so the electromagnetic flowmeter generally adopts an alternating magnetic field and is generated by the excitation of a 50HZ power frequency power source.
Measuring catheter: its function is to let the conductive liquid to be tested pass. In order to make the magnetic flux diverted or short-circuited when the magnetic flux passes through the measuring catheter, the measuring catheter must be made of non-magnetic, low electrical conductivity, low thermal conductivity and mechanical strength. Non-magnetic stainless steel, FRP, high strength can be used. Plastic, aluminum, etc.
Working principle and sensor parameters of ultrasonic external clamp flow meter
External clip type ultrasonic flow meter external clip type ultrasonic flow meter series measuring full flow of liquid flow meter, optional explosion-proof and intrinsically safe type, explosion-proof grade Exd[ib]IIBT6, and optional intrinsically safe remote control, explosion-proof grade ExibIIBT6, measurable medium: drinking water, river water, sea water, ground water cooling water, high temperature water, sewage, lubricating oil, diesel, fuel chemical liquid, other homogeneous fluid.
External clip sensor parameters:
The external clip type sensor is an installation method in which the sensor is directly bundled on the outer surface of the pipe to be measured to realize flow measurement, and has a diameter independent of the pipe diameter.
Simple installation, no need to stop production, no pressure loss and so on.
Third, gas turbine flowmeter daily inspection
1. Look at the display: Observe the total amount, instantaneous, temperature, pressure, and battery power. If the abnormality is reported in time.
2. Listen to the running sound of the flowmeter. (Method) The flowmeter should have no noise when it is running normally. If there is a large abnormal noise, it can be judged as damage to the movement of the movement (for example). It needs to be reported and processed in time.
3, data comparison, the output of the first 3 to 5 days before the statistics, if there is no major change, the flowmeter runs normally, such as the successive decrement phenomenon, the gas turbine flowmeter impeller (shaft, shaft core), filtration There is a problem with the device. Need to be reported in time for processing
4, gas turbine flowmeter filling lubricant
(1) Instrument refueling is an important part of the daily maintenance work of the instrument. Lubricating oil is mainly for instrument bearing; lubricating oil can protect the instrument bearing, reduce friction, reduce wear and prolong the operation cycle of the instrument;
(2) Regularly adding lubricating oil can effectively reduce the maintenance rate of the instrument and reduce the maintenance cost;
(3) Ensure the measurement accuracy of the flowmeter. "Measuring is counting money." why would you said this? The management of production and operation accounting, energy saving, cost accounting, etc. is an important production link of enterprise management. If the measurement is not in place, it will affect the economic efficiency and reputation of the enterprise. Therefore, the daily maintenance of the instrument is very important.