The lining material should be selected according to the corrosiveness, wear and temperature of the tested medium:
First, natural rubber (soft rubber)
1, better elasticity, wear resistance and tearing force
2, resistant to general weak acid, weak alkali corrosion
3. Water and sewage
Second, acid-resistant rubber (hard rubber)
It can withstand the corrosion of hydrochloric acid, acetic acid, oxalic acid, ammonia water, phosphoric acid and 50% sulfuric acid, sodium hydroxide and potassium hydroxide at normal temperature, but it is not resistant to corrosion by strong oxidants.
It can measure general acid, alkali and salt solutions.
Third, neoprene (Neoprene)
1, excellent elasticity, high tearing force, good wear resistance
2, resistant to general low concentration of acid and alkali, salt solution corrosion, but not resistant to oxidation of the medium, and temperature requirements <80 ° C;
It can measure water, sewage, mud and pulp.
Fourth, Polyurethane (Polyurethane)
1, excellent wear resistance
2, poor acid and alkali resistance, temperature requirements <40 ° C
Measure medium-strongly worn coal slurry, mud and slurry
Five, polytetrafluoroethylene (PTFE)
1. Boiling hydrochloric acid, sulfuric acid, nitric acid, aqua regia, concentrated alkali and various organic solvents
2, good wear resistance, poor bonding performance, temperature requirements -80 ~ +180 ° C;
Measuring concentration, concentrated alkali strong corrosive solution and sanitary medium
Advantages of turbine flow meters
The pressure loss is small, and the impeller has anti-corrosion function;
1.Adopt advanced ultra-low power single-chip microcomputer technology, the whole machine has strong functions, low power consumption and superior performance.
2.Intelligent flow display with nonlinear accuracy compensation. Correction formula accuracy is better than ±0.02%
3.The meter factor can be set online by button and can be displayed on the LCD screen. The LCD screen is intuitive and clear, and the reliability is strong.
4.EEPROM is used to accumulate the flow and the meter coefficient is powered down. The protection time is more than 10 years.
5.Using high-performance MCU central processing unit to complete data acquisition and processing display output, cumulative flow instantaneous flow
6.The screen displays a convenient man-machine interface for data transfer in standard 485 format.
7.Fully hard alloy (tungsten carbide) shielded cantilever beam structure bearing, integrating rotary bearing and pressure bearing, greatly improving bearing life and working in medium with a small amount of sediment and dirt.
8.It adopts 1Cr18Ni9Ti all stainless steel structure (the turbine adopts 2Cr13) and has good anti-corrosion performance.
9.Easy to maintain, self-rectifying structure, small and light, simple structure, which can be combined in a short time.
10.Internal cleaning is simple.
11.Strong anti-magnetic interference and vibration capability, reliable performance and long life
12.The lower limit flow rate is low and the measuring range is wide.
13.The on-site display type LCD screen is clear and intuitive, with low power consumption. The 3V lithium battery can be operated continuously for more than 5 years.
Corrosion resistant, suitable for acid and alkali solutions
Precautions before gas turbine flowmeter is enabled
Keywords: gas turbine flowmeter, Tianxin gas turbine flowmeter, gas gas turbine flowmeter
First, the precautions before the gas turbine flowmeter is enabled
In order to ensure trouble-free, safe start-up and commissioning of the ultrasonic flowmeter, the following pre-checks must be carried out:
1. Before starting the flowmeter officially, be sure to handle each leak point;
2. Check the power supply voltage on the terminal block and the corresponding positive and negative polarity;
3. Check the polarity of the frequency output lead connected to the RS485 converter;
4. Check whether the gas turbine flowmeter and temperature/pressure transmitter display are correct and reliable, including the installation direction and verticality of the flowmeter; the sealing performance of all joints;
5, wiring inspection: should be in accordance with the system wiring diagram provided by Jiangsu Tianxin Instrument Co., Ltd., check all wiring is correct.
Second, the gas turbine flowmeter projection operation steps
1. Open the bypass shut-off valve;
2. Open the flow upstream shutoff valve;
3. Slowly open the downstream shutoff valve of the flowmeter;
4. Slowly close the bypass shutoff valve.
Third, the gas turbine flowmeter stop table operation steps
1. Open the bypass shut-off valve;
2. Turn off the downstream shutoff valve of the flowmeter;
3. Close the flow upstream shutoff valve;
Fourth, gas turbine flowmeter use precautions
1. The newly installed or repaired pipeline must be purged. When purging the metering line, the flow meter must be removed and the corresponding short section replaced the flow meter for purging.
2. When the gas turbine flowmeter pipeline is put into production, it should be slowly boosted and gradually increase the flow rate. When the production is stopped, it should be slowly depressurized.
3. Check the sound of the gas turbine flowmeter or the vibration of the casing during operation to determine whether the turbine blades and bearings are working properly. At low flow rates, attention should be paid to the change in sound, and the vibration of the casing is observed at high flow rates.