The lining material should be selected according to the corrosiveness, wear and temperature of the tested medium:
First, natural rubber (soft rubber)
1, better elasticity, wear resistance and tearing force
2, resistant to general weak acid, weak alkali corrosion
3. Water and sewage
Second, acid-resistant rubber (hard rubber)
It can withstand the corrosion of hydrochloric acid, acetic acid, oxalic acid, ammonia water, phosphoric acid and 50% sulfuric acid, sodium hydroxide and potassium hydroxide at normal temperature, but it is not resistant to corrosion by strong oxidants.
It can measure general acid, alkali and salt solutions.
Third, neoprene (Neoprene)
1, excellent elasticity, high tearing force, good wear resistance
2, resistant to general low concentration of acid and alkali, salt solution corrosion, but not resistant to oxidation of the medium, and temperature requirements <80 ° C;
It can measure water, sewage, mud and pulp.
Fourth, Polyurethane (Polyurethane)
1, excellent wear resistance
2, poor acid and alkali resistance, temperature requirements <40 ° C
Measure medium-strongly worn coal slurry, mud and slurry
Five, polytetrafluoroethylene (PTFE)
1. Boiling hydrochloric acid, sulfuric acid, nitric acid, aqua regia, concentrated alkali and various organic solvents
2, good wear resistance, poor bonding performance, temperature requirements -80 ~ +180 ° C;
Measuring concentration, concentrated alkali strong corrosive solution and sanitary medium
Flowmeter product features:
Using intelligent judgment, the measurement correction setting is not used, and the air traffic control alarm and electrode detection application are more convenient;
Advanced "rough error handling" technology, which can remove fluids such as slurry to measure sharp disturbances, reduce output runout, maintain high precision measurement and make output more stable;
With a fluid density setting, it can display mass flow;
Constant current excitation current range, 125mA, 250mA optional, can be used with different manufacturers, different types of electromagnetic flow sensors;
Control function with remote reset of the totalizer, with contact signal input for starting and stopping accumulation, suitable for total inspection and batch processing applications;
With self-test and self-diagnosis function;
Advanced non-volatile memory for higher circuit reliability and effective protection of setup and measurement parameters;
The meter can be equipped with an unpowered clock and memory for recording the power down time, power-on time and power-down time;
The meter has an optional hour recording function that can store flow and electrode resistance measurements for more than 30 days.
The new keyboard processing method avoids the keyboard operation affecting the measurement, and can enter and return the operation menu to make the parameter setting more convenient;
The total display uses 10-bit decimal 9999999999 full-value carry, which solves the practice of double-word full value 4294967285 (hexadecimal FFFF) carry-in habit;
With wireless transmission, the mesh network is organized by the wireless HART protocol.
Third, gas turbine flowmeter daily inspection
1. Look at the display: Observe the total amount, instantaneous, temperature, pressure, and battery power. If the abnormality is reported in time.
2. Listen to the running sound of the flowmeter. (Method) The flowmeter should have no noise when it is running normally. If there is a large abnormal noise, it can be judged as damage to the movement of the movement (for example). It needs to be reported and processed in time.
3, data comparison, the output of the first 3 to 5 days before the statistics, if there is no major change, the flowmeter runs normally, such as the successive decrement phenomenon, the gas turbine flowmeter impeller (shaft, shaft core), filtration There is a problem with the device. Need to be reported in time for processing
4, gas turbine flowmeter filling lubricant
(1) Instrument refueling is an important part of the daily maintenance work of the instrument. Lubricating oil is mainly for instrument bearing; lubricating oil can protect the instrument bearing, reduce friction, reduce wear and prolong the operation cycle of the instrument;
(2) Regularly adding lubricating oil can effectively reduce the maintenance rate of the instrument and reduce the maintenance cost;
(3) Ensure the measurement accuracy of the flowmeter. "Measuring is counting money." why would you said this? The management of production and operation accounting, energy saving, cost accounting, etc. is an important production link of enterprise management. If the measurement is not in place, it will affect the economic efficiency and reputation of the enterprise. Therefore, the daily maintenance of the instrument is very important.