The lining material should be selected according to the corrosiveness, wear and temperature of the tested medium:
First, natural rubber (soft rubber)
1, better elasticity, wear resistance and tearing force
2, resistant to general weak acid, weak alkali corrosion
3. Water and sewage
Second, acid-resistant rubber (hard rubber)
It can withstand the corrosion of hydrochloric acid, acetic acid, oxalic acid, ammonia water, phosphoric acid and 50% sulfuric acid, sodium hydroxide and potassium hydroxide at normal temperature, but it is not resistant to corrosion by strong oxidants.
It can measure general acid, alkali and salt solutions.
Third, neoprene (Neoprene)
1, excellent elasticity, high tearing force, good wear resistance
2, resistant to general low concentration of acid and alkali, salt solution corrosion, but not resistant to oxidation of the medium, and temperature requirements <80 ° C;
It can measure water, sewage, mud and pulp.
Fourth, Polyurethane (Polyurethane)
1, excellent wear resistance
2, poor acid and alkali resistance, temperature requirements <40 ° C
Measure medium-strongly worn coal slurry, mud and slurry
Five, polytetrafluoroethylene (PTFE)
1. Boiling hydrochloric acid, sulfuric acid, nitric acid, aqua regia, concentrated alkali and various organic solvents
2, good wear resistance, poor bonding performance, temperature requirements -80 ~ +180 ° C;
Measuring concentration, concentrated alkali strong corrosive solution and sanitary medium
The main technical parameters of ultrasonic external clamp flowmeter:
Measurement accuracy: better than 1%
Repeatability: better than 0.2%
Measurement period: 500ms (2 times per second, 128 sets of data collected per cycle)
Working power: 220VAC/8~36VDC
Maximum flow rate: 64m / s (flow rate resolution 0.001m / s)
Display: 2 × 10 Chinese characters backlit LCD can display instantaneous flow and positive, negative, net cumulative flow, flow rate, etc.
Operation: 4×4 touch keyboard (F4 main unit magnetic 4 button) operation
Signal input: 3 channels of 4-20mA analog input, accuracy 0.1%, can input pressure, liquid level, temperature and other signals
2-way three-wire PT100 platinum resistor
Signal output: 1 isolated RS485 output
1 way 4-20mA or 0-20mA output
1 channel isolated OCT (programmable between pulse width 6~1000ms, default 200ms)
1 relay output (pulse width 200ms)
Data storage: Optional built-in data storage (SD card) can store time, instantaneous flow, accumulated flow, signal status, etc., through special software, can import data into computer for statistical and management
Communication protocol: MODBUS protocol, M-BUS protocol, FUJI extension protocol, and compatible with communication protocols of similar products of other domestic manufacturers
Other functions: automatic memory before 512 days, the first 128 months, the first 10 years of positive / negative / net cumulative flow
Automatically remember the first 30 times of power-on and power-off time and flow rate, and can realize automatic or manual addition of traffic, which can be read by MODBUS protocol.
Programmable batch (quantitative) controller, fault self-diagnosis function
Software upgrade via code file transmitted via E-mail
Protection level: sensor IP68, F4 host IP68, remaining host IP65
Explosion-proof grade: EXdIIBT4 (JN-100F2 type)
Steam vortex flowmeter measurement requirements
Keywords: steam vortex flowmeter, saturated steam vortex flowmeter, superheated steam vortex flowmeter
What should I pay attention to when measuring vortex flowmeter? What is the best way to measure steam installation by vortex flowmeter? Can vortex flowmeters be used for steam metering? The vortex flowmeters are evaluated for their measurement performance under single-phase flow conditions. There is currently no single-phase flowmeter for measuring the system variation of two-phase flow. However, the two-phase flow exists objectively, and its influence on flow measurement is unavoidable. Despite the difficulties, people are still trying to study the mechanism of its influence on flow measurement, and take corresponding countermeasures to improve the accuracy of flow measurement.
First, steam flow measurement difficulties
During long-distance transportation, dry steam will partially condense due to heat loss, resulting in reduced steam dryness and become wet steam. The gas-liquid two-phase flow structure in the horizontal pipe is related to the gas-liquid two-phase volume ratio and the flow velocity. In the steam pipe, since the volume ratio of the condensed water in the wet steam is small, the drain pipe drawn from the bottom of the horizontal pipe is made. , can receive a good hydrophobic effect. When the flow rate is particularly high, it will also behave as a circular flow, that is, there is a liquid film on the pipe wall, and the central part of the pipe is a gas core with droplets. Due to the influence of gravity during horizontal flow, the lower liquid film is higher than the upper pipe. The thickness of the wall, in the vertical ascending pipeline, the basic structure of the gas-liquid two-phase flow has a fine bubble flow structure, a bullet-like flow structure, a block flow structure, a ring-shaped flow structure with fibers, and an annular flow structure.