1. Requirements for straight pipe sections
Inlet/outlet straight pipe section: inlet should be ≥10×DN; outlet should be ≥5×DN
2. Grounding point requirement
In order to make the instrument work reliably, improve the measurement accuracy, free from external parasitic potential / the sensor should have good grounding, the grounding resistance is less than 10. (If the metal pipe is well grounded, no special grounding device is required) 3.3 Pair installation Location requirements as shown
Insert the electromagnetic flowmeter according to the situation of the pipeline in the field. If the flowmeter without the ball valve is installed, it should be installed on the pipeline without overpressure (that is, the flowmeter without the ball valve can be installed without pressure), and the hole is opened in the pipeline. Diameter 50, ready to weld the connecting welded pipe to the opening of the pipe; for occasions requiring constant flow loading or unloading or no media spillage, a ball valve must be installed, that is, a plug-in electromagnetic flowmeter with a ball valve structure is selected; With a diameter of 50, it is ready to weld the connecting welded pipe to the opening of the pipe.
Measuring range : Recommended range: 0.5m/s to 10m/s continuously adjustable. Maximum use range: continuously adjustable from 0.2m/s to 15m/s
Signal output: 1, the switch quantity can be set to: pulse output (up to 1000HZ); high/low flow alarm; empty pipe alarm; flow direction indication;
Fault alarm; 2, current output: 4-20mA output
Configuration method: 1. Field configuration through three manual keys. 2. Field configuration via remote control. 3. Perform on-site configuration through the handheld communicator.
Memory: The memory that the EEPROM does not disappear, no battery saving.
Ultrasonic flowmeter performance characteristics
Measuring liquid: single fluid such as water, seawater, alkali, etc., turbidity less than 10000ppm, particle size less than 1mm
Pipe material: Ultrasonic penetrating full pipe such as steel, cast iron, PVC pipe, dense texture, no serious corrosion and scaling, allowing thin layer of dense lining
Pipe lining material: rubber, epoxy asphalt, fiberglass, mortar, etc., or unlined
Pipe diameter range: DN15mm~6000mm, plug-in sensor for pipe diameter DN25mm~6000mm
Flow rate range: -32m/s~0~+32m/s, bidirectional flow
Accuracy: ±1.0% of displayed value (under standard conditions). Configure plug-in sensor accuracy up to 0.5%
Signal output: 4-20mA or 0-20mA RS-485
Sensor: It is divided into external clip type and plug-in type. The standard configuration is clip-on. The maximum operating pressure of the plug-in sensor is 3 MPa.
Straight pipe length: upstream >10D, downstream >5D from pump outlet >30D (D: pipe diameter)
Display mode: 2*20 character LCD display with LED backlight
Protection level: IP65
Explosion-proof grade: ExdIIBT4
Fixed ultrasonic flowmeter power supply: AC110V, AC220V, DC8~36V or AC7~30V
Ambient temperature / humidity: Host -20 ° C ~ +80 ° C; sensor -20 ° C ~ +80 ° C or -20 ° C ~ +160 ° C optional, the sensor can be immersed in water, water depth less than 2m.
Compressed air flowmeter installation requirements
Keywords: compressed air flow meter, compressed air vortex flowmeter, air flowmeter
First, the compressed air vortex flowmeter installation requirements
1. The upstream of the compressed air vortex flowmeter should avoid installing the regulating valve or the semi-opening valve. The regulating valve or the semi-opening valve is installed after the downstream 8DN of the sensor.
2. The straight pipe section where the flowmeter is installed should be as close as possible to the sensor diameter. If it is not consistent, a pipe diameter slightly larger than the sensor diameter should be used, and the error should be ≤3% and not more than 5mm.
3. When the measured medium contains more impurities, the filter should be installed outside the length required for the straight pipe section upstream of the sensor.
4, the sensor should be avoided on the pipeline with mechanical vibration, and try to avoid strong electromagnetic field interference. When vibration cannot be avoided, consider adding a bracket to the straight pipe section about 2DN before and after the sensor.