The electromagnetic flowmeter has a large measuring range, usually 20:1~50:1, and the optional flow range is wide; the diameter of the electromagnetic flowmeter is wider than other types of flowmeters, from a few millimeters to 3 meters; the positive and negative two-way flow can be measured. The pulsating flow rate can also be measured as long as the pulsation frequency is much lower than the excitation frequency; the meter output is linear in nature; it is easy to select the material type of the fluid contact member, and can be applied to corrosive fluids and the like. Since the electromagnetic flowmeter measures much more with suspended solids or dirt than other flow meters, the probability of failure of the inner wall adhesion layer is relatively high. If the electrical conductivity of the adhesion layer is close to the liquid conductivity, the instrument can output the signal normally, but only change the flow area to form a hidden fault of the measurement error; if it is a high conductivity adhesion layer, the electromotive force between the electrodes will be short-circuited; if it is an insulating adhesion layer The surface of the electrode is insulated to disconnect the measuring circuit. The latter two phenomena will make the meter inoperable.
Advantages of turbine flow meters
The pressure loss is small, and the impeller has anti-corrosion function;
1.Adopt advanced ultra-low power single-chip microcomputer technology, the whole machine has strong functions, low power consumption and superior performance.
2.Intelligent flow display with nonlinear accuracy compensation. Correction formula accuracy is better than ±0.02%
3.The meter factor can be set online by button and can be displayed on the LCD screen. The LCD screen is intuitive and clear, and the reliability is strong.
4.EEPROM is used to accumulate the flow and the meter coefficient is powered down. The protection time is more than 10 years.
5.Using high-performance MCU central processing unit to complete data acquisition and processing display output, cumulative flow instantaneous flow
6.The screen displays a convenient man-machine interface for data transfer in standard 485 format.
7.Fully hard alloy (tungsten carbide) shielded cantilever beam structure bearing, integrating rotary bearing and pressure bearing, greatly improving bearing life and working in medium with a small amount of sediment and dirt.
8.It adopts 1Cr18Ni9Ti all stainless steel structure (the turbine adopts 2Cr13) and has good anti-corrosion performance.
9.Easy to maintain, self-rectifying structure, small and light, simple structure, which can be combined in a short time.
10.Internal cleaning is simple.
11.Strong anti-magnetic interference and vibration capability, reliable performance and long life
12.The lower limit flow rate is low and the measuring range is wide.
13.The on-site display type LCD screen is clear and intuitive, with low power consumption. The 3V lithium battery can be operated continuously for more than 5 years.
Corrosion resistant, suitable for acid and alkali solutions
Third, gas turbine flowmeter daily inspection
1. Look at the display: Observe the total amount, instantaneous, temperature, pressure, and battery power. If the abnormality is reported in time.
2. Listen to the running sound of the flowmeter. (Method) The flowmeter should have no noise when it is running normally. If there is a large abnormal noise, it can be judged as damage to the movement of the movement (for example). It needs to be reported and processed in time.
3, data comparison, the output of the first 3 to 5 days before the statistics, if there is no major change, the flowmeter runs normally, such as the successive decrement phenomenon, the gas turbine flowmeter impeller (shaft, shaft core), filtration There is a problem with the device. Need to be reported in time for processing
4, gas turbine flowmeter filling lubricant
(1) Instrument refueling is an important part of the daily maintenance work of the instrument. Lubricating oil is mainly for instrument bearing; lubricating oil can protect the instrument bearing, reduce friction, reduce wear and prolong the operation cycle of the instrument;
(2) Regularly adding lubricating oil can effectively reduce the maintenance rate of the instrument and reduce the maintenance cost;
(3) Ensure the measurement accuracy of the flowmeter. "Measuring is counting money." why would you said this? The management of production and operation accounting, energy saving, cost accounting, etc. is an important production link of enterprise management. If the measurement is not in place, it will affect the economic efficiency and reputation of the enterprise. Therefore, the daily maintenance of the instrument is very important.