The electromagnetic flowmeter has a large measuring range, usually 20:1~50:1, and the optional flow range is wide; the diameter of the electromagnetic flowmeter is wider than other types of flowmeters, from a few millimeters to 3 meters; the positive and negative two-way flow can be measured. The pulsating flow rate can also be measured as long as the pulsation frequency is much lower than the excitation frequency; the meter output is linear in nature; it is easy to select the material type of the fluid contact member, and can be applied to corrosive fluids and the like. Since the electromagnetic flowmeter measures much more with suspended solids or dirt than other flow meters, the probability of failure of the inner wall adhesion layer is relatively high. If the electrical conductivity of the adhesion layer is close to the liquid conductivity, the instrument can output the signal normally, but only change the flow area to form a hidden fault of the measurement error; if it is a high conductivity adhesion layer, the electromotive force between the electrodes will be short-circuited; if it is an insulating adhesion layer The surface of the electrode is insulated to disconnect the measuring circuit. The latter two phenomena will make the meter inoperable.
Flowmeter product features:
Using intelligent judgment, the measurement correction setting is not used, and the air traffic control alarm and electrode detection application are more convenient;
Advanced "rough error handling" technology, which can remove fluids such as slurry to measure sharp disturbances, reduce output runout, maintain high precision measurement and make output more stable;
With a fluid density setting, it can display mass flow;
Constant current excitation current range, 125mA, 250mA optional, can be used with different manufacturers, different types of electromagnetic flow sensors;
Control function with remote reset of the totalizer, with contact signal input for starting and stopping accumulation, suitable for total inspection and batch processing applications;
With self-test and self-diagnosis function;
Advanced non-volatile memory for higher circuit reliability and effective protection of setup and measurement parameters;
The meter can be equipped with an unpowered clock and memory for recording the power down time, power-on time and power-down time;
The meter has an optional hour recording function that can store flow and electrode resistance measurements for more than 30 days.
The new keyboard processing method avoids the keyboard operation affecting the measurement, and can enter and return the operation menu to make the parameter setting more convenient;
The total display uses 10-bit decimal 9999999999 full-value carry, which solves the practice of double-word full value 4294967285 (hexadecimal FFFF) carry-in habit;
With wireless transmission, the mesh network is organized by the wireless HART protocol.
Precautions before gas turbine flowmeter is enabled
Keywords: gas turbine flowmeter, Tianxin gas turbine flowmeter, gas gas turbine flowmeter
First, the precautions before the gas turbine flowmeter is enabled
In order to ensure trouble-free, safe start-up and commissioning of the ultrasonic flowmeter, the following pre-checks must be carried out:
1. Before starting the flowmeter officially, be sure to handle each leak point;
2. Check the power supply voltage on the terminal block and the corresponding positive and negative polarity;
3. Check the polarity of the frequency output lead connected to the RS485 converter;
4. Check whether the gas turbine flowmeter and temperature/pressure transmitter display are correct and reliable, including the installation direction and verticality of the flowmeter; the sealing performance of all joints;
5, wiring inspection: should be in accordance with the system wiring diagram provided by Jiangsu Tianxin Instrument Co., Ltd., check all wiring is correct.
Second, the gas turbine flowmeter projection operation steps
1. Open the bypass shut-off valve;
2. Open the flow upstream shutoff valve;
3. Slowly open the downstream shutoff valve of the flowmeter;
4. Slowly close the bypass shutoff valve.
Third, the gas turbine flowmeter stop table operation steps
1. Open the bypass shut-off valve;
2. Turn off the downstream shutoff valve of the flowmeter;
3. Close the flow upstream shutoff valve;
Fourth, gas turbine flowmeter use precautions
1. The newly installed or repaired pipeline must be purged. When purging the metering line, the flow meter must be removed and the corresponding short section replaced the flow meter for purging.
2. When the gas turbine flowmeter pipeline is put into production, it should be slowly boosted and gradually increase the flow rate. When the production is stopped, it should be slowly depressurized.
3. Check the sound of the gas turbine flowmeter or the vibration of the casing during operation to determine whether the turbine blades and bearings are working properly. At low flow rates, attention should be paid to the change in sound, and the vibration of the casing is observed at high flow rates.