Application of electromagnetic flowmeter
Because electromagnetic flowmeter has its unique advantages, it is widely used in chemical, chemical fiber, food, paper, sugar, water supply and drainage, environmental protection, water conservancy, steel, petroleum, pharmaceutical and other industrial fields. Various acid and alkali salt solutions, mud, pulp, pulp, coal water slurry, corn syrup, fiber pulp, lime slurry, water supply and drainage, brine, hydrogen peroxide, beer, wort, various beverages, black liquor, green liquor, etc. The volumetric flow of the medium.
The main technical parameters
Nominal diameter series DN (mm)
Pipeline PTFE lining:
Pipeline rubber lining:
Special specifications can be customized
Flow direction: current output
Positive, negative, net flow
Range ratio: 150:1
Repeatability error: ±0.1 of the measured value
Accuracy level: pipeline type: 0.5 level, 1.0 level
Measured medium temperature
Ordinary rubber lining: -20 ~ +60 ° C
High temperature rubber lining: -20 ~ +90 ° C
PTFE lining: -30 ~ +100 ° C
High temperature PTFE lining: -30 ~ +180 ° C
Rated operating voltage
Pipe type: DN6-DN100≤1.6Mpa, DN125-DN225≤1.0Mpa high pressure can be customized
Conductivity of the measured fluid ≥ 5us / cm (integrated)
Most water-based media have a conductivity of 200-800 us/cm, and electromagnetic flow meters can be used to measure the flow rate.
Load resistance: 0 ~ 10mA, 0 ~ 1.5kΩ
4 to 20 mA, 0 to 750 Ω
Digital frequency output
The upper limit of the output frequency can be set to open the bidirectional output of the open collector of the transistor with optical isolation from 1 to 5000HZ. External ≤35V, the maximum current of the collector is 25mA when conducting
Straight pipe length
Pipeline type: upstream ≥5DN, downstream ≥2DN
The flowmeter and the pipe are flanged, and the flange connection size should meet the requirements of GB11988.
Explosion-proof mark: mdllBT4
Ambient temperature: -10 ° C ~ +55 ° C
Relative temperature: 5% to 90%
Total power consumption: less than 15W
Gas turbine flowmeter and screwing vortex flowmeter measurement difference
Keywords: gas turbine flowmeter, Tianxin gas turbine flowmeter, gas gas turbine flowmeter
First, the characteristics of gas turbine flowmeter and spiral vortex flowmeter
Gas turbine flowmeter and screwing vortex flowmeter are the main gas metering instruments in China. In foreign countries, almost no one uses gas to swirl into the vortex flow meter. At present, there are certain applications in the domestic vortex flowmeter. In terms of standard conditions. The international standard ISO9551 promulgated by the International Organization for Standardization and the international recommendations R6, R32 promulgated by the International Organization of Legal Metrology (OIML) have corresponding national standards and regulations in China and developed countries. The above documents guarantee the authority of gas turbine flowmeters in gas metering. Sex. Gas-introducing vortex flowmeters have not yet established relevant standards. It is difficult to guarantee the authority of the flow meter in gas metering. At present, in domestic flow meters for gas measurement, gas turbine flowmeters dominate, and the vortex flowmeters also have a certain market. The use of the two is similar, and users often have doubts about the choice of the two.
Steam vortex flowmeter measurement requirements
Keywords: steam vortex flowmeter, saturated steam vortex flowmeter, superheated steam vortex flowmeter
What should I pay attention to when measuring vortex flowmeter? What is the best way to measure steam installation by vortex flowmeter? Can vortex flowmeters be used for steam metering? The vortex flowmeters are evaluated for their measurement performance under single-phase flow conditions. There is currently no single-phase flowmeter for measuring the system variation of two-phase flow. However, the two-phase flow exists objectively, and its influence on flow measurement is unavoidable. Despite the difficulties, people are still trying to study the mechanism of its influence on flow measurement, and take corresponding countermeasures to improve the accuracy of flow measurement.
First, steam flow measurement difficulties
During long-distance transportation, dry steam will partially condense due to heat loss, resulting in reduced steam dryness and become wet steam. The gas-liquid two-phase flow structure in the horizontal pipe is related to the gas-liquid two-phase volume ratio and the flow velocity. In the steam pipe, since the volume ratio of the condensed water in the wet steam is small, the drain pipe drawn from the bottom of the horizontal pipe is made. , can receive a good hydrophobic effect. When the flow rate is particularly high, it will also behave as a circular flow, that is, there is a liquid film on the pipe wall, and the central part of the pipe is a gas core with droplets. Due to the influence of gravity during horizontal flow, the lower liquid film is higher than the upper pipe. The thickness of the wall, in the vertical ascending pipeline, the basic structure of the gas-liquid two-phase flow has a fine bubble flow structure, a bullet-like flow structure, a block flow structure, a ring-shaped flow structure with fibers, and an annular flow structure.