Ultrasonic flowmeter features
The unique signal digitization processing technology makes the meter measurement signal more stable, anti-interference ability and more accurate measurement.
No mechanical transmission parts are not easily damaged, maintenance-free and have a long service life.
The circuit is more optimized, with high integration, low power consumption and high reliability.
Intelligent standard signal output, friendly man-machine interface, multiple secondary signal output, for you to choose.
Pipe-section small pipe diameter measurement is economical and convenient, and the measurement accuracy is high.
Detailed installation of ultrasonic flowmeter
Ultrasonic flowmeters should be aware of the site prior to installation, including:
1. What is the distance from the host at the installation of the sensor;
2. Pipe material, pipe wall thickness and pipe diameter;
3. Years of pipeline;
4. The type of fluid, whether it contains impurities, bubbles and whether it is full;
5. Fluid temperature;
6. Whether there is interference source at the installation site (such as frequency conversion, strong magnetic field, etc.);
7. Four seasons temperature at the host place;
8. Whether the power supply voltage used is stable;
9. Do you need remote signals and types;
According to the site conditions provided above, the manufacturer can configure the site conditions and, if necessary, special models.
According to different output signals, these products can be divided into LWGY-□N type and LWGY-□A type.
LWGY-□N type sensor: 12～24VDC power supply, three-wire pulse output, high level ≥8V, low level ≤0.8V; signal transmission distance ≤1000m;
LWGY-□A type transmitter: 24VDC power supply, 2-wire 4-20mA output, signal transmission distance ≤1000.
This type of turbine flow products are divided into basic type and explosion-proof type (ExdIIBT6).
2, intelligent integrated turbine flowmeter
Intelligent integrated turbine flowmeter adopts advanced ultra-low-power single-chip microcomputer technology to develop a new intelligent instrument integrating turbine flow sensor and display integration. It adopts double-row liquid crystal field display, with compact structure, intuitive reading and reliability. High, free from external power interference, lightning strike, low cost and other obvious advantages. The instrument has three-point correction of the meter coefficient, the intelligent compensation meter coefficient is nonlinear, and field correction can be performed. The high-definition LCD displays both instantaneous flow (4 significant digits) and cumulative flow (8 significant digits with clear function). All valid data is not lost after 10 years of power failure. These turbine flowmeters are explosion-proof products with an explosion-proof rating of ExdIIBT6.
However, since the volume of condensed water in the wet steam is relatively small, the flow of excess water in the ascending pipe often appears as an annular flow structure, but when the amount of water is particularly large, it also appears as a ring with fibers. Flow structure. Among them, the fibrous fluid is actually a condensed water. Flow in a vertically descending pipe: In a vertically descending pipe, the structure of the gas-liquid two-phase flow is similar to that of a vertically ascending flow, but differs not only in the opposite flow direction but also in the case of the same average flow velocity. The flow rate of the liquid in the vertical drop pipe is much faster than the flow rate of the liquid in the vertical rise pipe.
Second, steam vortex flowmeter measurement considerations
Vortex flowmeters measure steam. Accurately arrange the steam traps accurately: People have long discovered that when the steam has more water, the vortex flowmeter will have a "leakage pulse" phenomenon, that is, in the case of a stable steam flow rate, the vortex flow It should have a stable pulse output proportional to the flow rate. But sometimes it is found that the output pulse of the meter is inexplicably less. The distribution of the output pulse recorded on the two-dimensional coordinates can also be clearly seen. The pulse that should be approximately evenly distributed is one less at a place. Pulses, in severe cases, are a lot less pulses, and in the worst case, there are no pulses at all. This may be related to the formation of a vortex column by impinging on a large volume of droplets having a non-uniform distribution on the vortex generator.