Use and promotion of electromagnetic flowmeter
The intelligent electromagnetic flowmeter is a fully intelligent electromagnetic flowmeter developed by our company using advanced technology at home and abroad. Its all-Chinese electromagnetic converter core adopts high-speed central processing unit, which has fast calculation speed, high precision and reliable measurement performance. The converter circuit design adopts the latest international technology, the input impedance is up to 1015 ohms, the common mode rejection ratio is better than 100db, and the external interference and 60Hz/50Hz interference suppression ability is better than 90db, which can measure the lower conductivity fluid medium flow. The sensor adopts a new non-uniform magnetic field technology and a special magnetic circuit structure. The magnetic field is stable and reliable, and the volume is greatly reduced, the weight is reduced, and the flowmeter has the characteristics of small size and light weight. Adhering to the enterprise spirit of "seeking truth, hard work, innovation and development", we insist on "customer first, technology first, high quality, first-class service", we will provide you with first-class products and first-class service.
Third, the spiral vortex flowmeter measurement characteristics
The spiral vortex flowmeter is a flowmeter that began to appear in the 1970s. Its working principle is: the gas that enters the gas swirling vortex flowmeter is first forced by the spiral spinner to accelerate the rotation to form a vortex, the center of the vortex. For the vortex core. The accelerated vortex enters the enlarged section and then decelerates sharply. The pressure rises to produce a recirculation. Under the action of the recirculation, the vortex core makes a spiral precession around the axis of the flowmeter. The flow rate Q can be derived by measuring the vortex precession frequency f by sensing the sensitive component.
Fourth, gas turbine flowmeter and spiral vortex flowmeter measurement difference
The gas turbine flowmeter has a small pressure loss and can be suitable for gas metering in low pressure conveying applications. Gas-injected vortex flowmeters have a slightly higher pressure loss, and gas metering in low-pressure delivery situations sometimes has problems.
Gas turbine flowmeters require high media cleanliness and can be damaged if used improperly. The gas swirling vortex flowmeter gas swirling vortex flowmeter requires no gas turbine flowmeter and is not damaged.
Gas turbine flowmeters are highly accurate. It is the most accurate of all flow meters. The gas spiraling vortex flowmeter is not as accurate as the former.
Precautions before gas turbine flowmeter is enabled
Keywords: gas turbine flowmeter, Tianxin gas turbine flowmeter, gas gas turbine flowmeter
First, the precautions before the gas turbine flowmeter is enabled
In order to ensure trouble-free, safe start-up and commissioning of the ultrasonic flowmeter, the following pre-checks must be carried out:
1. Before starting the flowmeter officially, be sure to handle each leak point;
2. Check the power supply voltage on the terminal block and the corresponding positive and negative polarity;
3. Check the polarity of the frequency output lead connected to the RS485 converter;
4. Check whether the gas turbine flowmeter and temperature/pressure transmitter display are correct and reliable, including the installation direction and verticality of the flowmeter; the sealing performance of all joints;
5, wiring inspection: should be in accordance with the system wiring diagram provided by Jiangsu Tianxin Instrument Co., Ltd., check all wiring is correct.
Second, the gas turbine flowmeter projection operation steps
1. Open the bypass shut-off valve;
2. Open the flow upstream shutoff valve;
3. Slowly open the downstream shutoff valve of the flowmeter;
4. Slowly close the bypass shutoff valve.
Third, the gas turbine flowmeter stop table operation steps
1. Open the bypass shut-off valve;
2. Turn off the downstream shutoff valve of the flowmeter;
3. Close the flow upstream shutoff valve;
Fourth, gas turbine flowmeter use precautions
1. The newly installed or repaired pipeline must be purged. When purging the metering line, the flow meter must be removed and the corresponding short section replaced the flow meter for purging.
2. When the gas turbine flowmeter pipeline is put into production, it should be slowly boosted and gradually increase the flow rate. When the production is stopped, it should be slowly depressurized.
3. Check the sound of the gas turbine flowmeter or the vibration of the casing during operation to determine whether the turbine blades and bearings are working properly. At low flow rates, attention should be paid to the change in sound, and the vibration of the casing is observed at high flow rates.