Use and promotion of electromagnetic flowmeter
The intelligent electromagnetic flowmeter is a fully intelligent electromagnetic flowmeter developed by our company using advanced technology at home and abroad. Its all-Chinese electromagnetic converter core adopts high-speed central processing unit, which has fast calculation speed, high precision and reliable measurement performance. The converter circuit design adopts the latest international technology, the input impedance is up to 1015 ohms, the common mode rejection ratio is better than 100db, and the external interference and 60Hz/50Hz interference suppression ability is better than 90db, which can measure the lower conductivity fluid medium flow. The sensor adopts a new non-uniform magnetic field technology and a special magnetic circuit structure. The magnetic field is stable and reliable, and the volume is greatly reduced, the weight is reduced, and the flowmeter has the characteristics of small size and light weight. Adhering to the enterprise spirit of "seeking truth, hard work, innovation and development", we insist on "customer first, technology first, high quality, first-class service", we will provide you with first-class products and first-class service.
Liquid turbine flow meter classification and function description
Keywords: liquid turbine flowmeter, diesel liquid turbine flowmeter, pure water liquid turbine flowmeter
First, the liquid turbine flowmeter overview
The liquid turbine flowmeter is the main type of impeller flow (flow rate) meter, and the impeller type flowmeter also has an anemometer and a water meter. The TUF consists of a sensor and a conversion display. The sensor uses a multi-blade rotor to sense the average flow rate of the fluid, thereby deriving the flow or total amount. The rotational speed (or number of revolutions) of the rotor can be detected mechanically, magnetically, and photoelectrically and displayed and transmitted by the reading device.
Second, liquid turbine flowmeter classification
According to the classification of instrument functions, SDLWGY series turbine flowmeters can be divided into 2 categories.
1, turbine flow sensor / transmitter
2, intelligent integrated turbine flowmeter
Third, the liquid turbine flowmeter function description
1, turbine flow sensor / transmitter
This type of turbine flow product itself does not have a live display function, only the flow signal is transmitted far. The flow signal can be divided into pulse signal or current signal (4-20mA); the instrument is low in price, high in integration and small in size, and is especially suitable for use with computer control systems such as secondary display, PLC, DCS.
Compressed air flowmeter installation requirements
Keywords: compressed air flow meter, compressed air vortex flowmeter, air flowmeter
First, the compressed air vortex flowmeter installation requirements
1. The upstream of the compressed air vortex flowmeter should avoid installing the regulating valve or the semi-opening valve. The regulating valve or the semi-opening valve is installed after the downstream 8DN of the sensor.
2. The straight pipe section where the flowmeter is installed should be as close as possible to the sensor diameter. If it is not consistent, a pipe diameter slightly larger than the sensor diameter should be used, and the error should be ≤3% and not more than 5mm.
3. When the measured medium contains more impurities, the filter should be installed outside the length required for the straight pipe section upstream of the sensor.
4, the sensor should be avoided on the pipeline with mechanical vibration, and try to avoid strong electromagnetic field interference. When vibration cannot be avoided, consider adding a bracket to the straight pipe section about 2DN before and after the sensor.