Electrode: Its function is to extract and induce a proportional induced inductive potential signal. The electrodes are typically made of non-magnetically conductive stainless steel and are required to be flush with the liner so that the fluid passes unimpeded. It should be installed in the vertical direction of the pipe to prevent deposits from accumulating on it and affecting the measurement accuracy.
Enclosure: Made of ferromagnetic material, it is the cover of the distribution system excitation coil and isolates the interference of external magnetic field.
Lining: A complete electrical insulation lining on the inside of the measuring tube and on the flange sealing surface. It directly contacts the liquid to be measured, and its function is to increase the corrosion resistance of the measuring catheter and prevent the induced potential from being short-circuited by the metal measuring tube wall. Most of the lining materials are PTFE plastics and ceramics that are resistant to corrosion, high temperature and wear.
Converter: The induced potential signal generated by the liquid flow is very weak and is greatly affected by various interference factors. The function of the converter is to amplify and convert the induced potential signal into a unified standard signal and suppress the main interference signal. Its task is to amplify the induced potential signal Ex detected by the electrode into a unified standard DC signal.
Plug-in electromagnetic flowmeter features
Corrosion-resistant full stainless steel material, ABS full mold production probe structure is simple, firm, no moving parts, long service life, reliable measurement, strong anti-interference ability, small size, light weight, easy installation, small maintenance range, large measurement range, temperature-free measurement The influence of changes in density, pressure, viscosity, conductivity, etc., zero pressure loss can be installed and disassembled under continuous flow, which brings convenience to the user's maintenance. The manufacturing cost and installation cost are lower than the general electromagnetic flowmeter. It is especially suitable for large and medium-diameter pipeline flow measurement. It adopts advanced processing technology, solid-state packaging, vibration resistance and long life. It makes the meter have good measurement accuracy and stability. The flowmeter not only has 4-20mA current signal output, but also has 1-2000HZ frequency output digital communication output: RS485, RS232, HART, Modbus.
Installation of plug-in electromagnetic flowmeter
The plug-in electromagnetic flowmeter sensor installation upstream straight pipe section is not less than 5D (D means pipe diameter), the downstream straight pipe section is not less than 2D:
1. Vertical installation. When the sensor is inserted into the pipeline, it should be less than 5° from the vertical diameter of the pipeline section. It is suitable for measuring the cleaning medium with small vibration of the pipeline.
2, inclined installation, the axis of the sensor and the axis of the pipe to be tested is 45 °, suitable for large pipe diameter and measuring the flow of liquid containing other impurities in the medium, the installation method is small, not suitable for entanglement.
There are two ways to insert the insertion point of the electromagnetic flowmeter sensor: one is inserted into the central axis of the pipe to be tested (ie 1/2 times the diameter of the pipe); the other is inserted into the inner wall of the pipe at 1/4 of the pipe.
1. Clean the welding slag and burrs of the base of the pipe to be tested.
2. Turn off the upstream flow control valve or use low pressure water supply.
3. Install the DN50 ball valve on the mounting base. Pay attention to the long cavity of the ball valve. Check whether the ball valve can be fully open and fully closed. Install the screw seat, compression nut and rubber sealing ring onto the ball valve and loosen the positioning nut. Insert the sensor into the rod through the ball valve and insert it into the pipe to be tested. At the same time, it should be noted that the direction of the sensor direction indicator rod should be consistent with the flow direction of the fluid. Ordering Information Please specify the name of the tested medium, working pressure, medium temperature, flow rate (flow rate) range, pipe diameter and the pipe network to be tested.