Ultrasonic flowmeter uses the electronic principle to measure the flow meter. Although it has its own verification procedure, because it has the same usage management mode as the tap water industry measurement, it puts its performance requirements, measurement accuracy, verification cycle, etc. into the cold water meter. In the meantime, it has become more standardized. Since then, the use of large-caliber ultrasonic flowmeters has normative documents in the water industry.
Ultrasonic flowmeter (ultrasonic water meter) Because the measurement principle is different from electromagnetic water meter, its accuracy is different. How to choose an ultrasonic flowmeter with superior performance and precision:
The user flow range is used to determine the type, diameter, temperature, medium, etc. of the ultrasonic flowmeter, especially considering the common flow rate and the starting flow rate. The purpose is to ensure the water supply quantity, accurately measure and reduce the leakage rate. When using the water meter, the user's water meter Commonly used traffic generally cannot exceed the common flow value specified by the product.
Different flow rate to determine the diameter of the flow meter can meet the measurement range.
Different user environments.
Also consider the working pressure, the structure of the flow certificate, the form of the converter (whether bidirectional flow, current, pulse, communication interface, data storage) and the length of the connecting cable.
The performance and accuracy of the ultrasonic flowmeter are selected.
Users can learn more about the peers who are already using the ultrasonic flowmeter, to see the effect of their use, the accuracy of the measurement.
At present, industrial flow measurement generally has problems of large diameter, large flow, small diameter, small flow, high temperature liquid, and gas measurement. This is because the general flowmeter will bring manufacturing and transportation as the diameter of the measuring pipe increases. The difficulty of the above, the advantages of increased cost, increased energy loss, and inconvenient installation, ultrasonic flowmeters can be avoided. The German FLEXIM ultrasonic flowmeter measures diameters from 6mm to 6500 mm, from a few meters wide open channels, culverts to 500m wide rivers.
Turbine flow meter applications
Turbine flow meters are widely used in the following measurement objects: petroleum, organic liquids, inorganic liquids, liquefied gases, natural gas, gas and cryogenic fluids. In the transshipment and gathering stations of foreign liquefied petroleum gas, refined oil and light crude oil, the first and last stations of large crude oil transmission pipelines use it for trade settlement. In Europe and the United States, the turbine flowmeter is the natural gas meter next to the orifice flowmeter. In the Netherlands alone, more than 2,600 gas turbine flowmeters of various sizes and pressures from 0.8 MPa to 6.5 MPa are used on the natural gas pipeline. Has become an excellent natural gas flow meter. Although the excellent metering characteristics of the turbine flow meter are favored by people, the impression is that the moving parts are short in use time, and they are inevitable in the selection. After people's unremitting efforts, it should be said that the situation has changed a lot. As the most versatile flowmeter, the turbine flowmeter has been developed into a multi-variety, full-series, multi-size mass production scale.
Installation of plug-in electromagnetic flowmeter
The plug-in electromagnetic flowmeter sensor installation upstream straight pipe section is not less than 5D (D means pipe diameter), the downstream straight pipe section is not less than 2D:
1. Vertical installation. When the sensor is inserted into the pipeline, it should be less than 5° from the vertical diameter of the pipeline section. It is suitable for measuring the cleaning medium with small vibration of the pipeline.
2, inclined installation, the axis of the sensor and the axis of the pipe to be tested is 45 °, suitable for large pipe diameter and measuring the flow of liquid containing other impurities in the medium, the installation method is small, not suitable for entanglement.
There are two ways to insert the insertion point of the electromagnetic flowmeter sensor: one is inserted into the central axis of the pipe to be tested (ie 1/2 times the diameter of the pipe); the other is inserted into the inner wall of the pipe at 1/4 of the pipe.
1. Clean the welding slag and burrs of the base of the pipe to be tested.
2. Turn off the upstream flow control valve or use low pressure water supply.
3. Install the DN50 ball valve on the mounting base. Pay attention to the long cavity of the ball valve. Check whether the ball valve can be fully open and fully closed. Install the screw seat, compression nut and rubber sealing ring onto the ball valve and loosen the positioning nut. Insert the sensor into the rod through the ball valve and insert it into the pipe to be tested. At the same time, it should be noted that the direction of the sensor direction indicator rod should be consistent with the flow direction of the fluid. Ordering Information Please specify the name of the tested medium, working pressure, medium temperature, flow rate (flow rate) range, pipe diameter and the pipe network to be tested.