Electromagnetic flowmeter features
Frequency programmable low frequency rectangular wave excitation improves stability of flow measurement and low power loss.
The new 16-bit ultra-low power microprocessor with FLASH memory has high integration, fast calculation speed and high calculation accuracy.
Full digital processing, strong anti-interference ability and reliable measurement.
Ultra-low EMI switching power supply, suitable for power supply voltage variation range, high efficiency, small temperature rise; good EMC performance.
Chinese and English menu operation, easy to use, easy to operate, easy to learn and understand.
High definition backlight wide temperature LCD display.
It can perform bidirectional flow measurement and bidirectional total accumulation; it has automatic range switching function, which can effectively improve the measurement accuracy of analog current and frequency output, especially suitable for occasions where the diurnal flow range changes greatly and needs to send control signals; flow measurement range Up to 1500:1.
There are three totalizers inside, which record and display the forward cumulative amount, the reverse cumulative amount and the accumulated difference integrated amount, which are convenient for fluid metering and custody transfer.
Provide isolated or non-isolated RS485/RS232C digital communication interface, and support fieldbus communication modes such as MODBUS, PROFIBUS-DP and HART.
Constant current source fluid resistance measurement can accurately measure the internal resistance of the electrode signal in the case of long-line transmission. It can be used not only to determine whether the fluid in the sensor is empty or not, but also to identify abnormal phenomena such as contamination and coverage of the electrode, and to provide cleaning for the user. Fault processing information such as electrodes.
In the case of small flow metering, the film surface ratio is the largest, the Roots flowmeter is second, and the turbine flowmeter is the smallest. The initial flow performance of the volumetric flowmeter should be superior to the velocity flowmeter. For industrial and commercial users with large total gas consumption and low heat load for each gas, the volumetric flowmeter, ie, the membrane meter and the Roots flowmeter, should be preferred to meet the flowmeter requirements of a single burner when used alone. It can also meet the measurement requirements of the maximum flow rate when all burners are used at the same time.
Not all industrial and commercial users can solve the measurement problem by selecting a volumetric flowmeter with a larger range. The volumetric ratio of the volumetric flowmeter is limited, and it is impossible to increase without limit. The selection of the flowmeter cannot be solved. This kind of user has a measurement problem. When the user only uses a single device for gas, or when operating at low flow rate, the initial flow rate and minimum flow rate of a single flow meter may not meet the minimum flow rate requirement. A single large flow meter cannot measure low gas consumption, which will result in Large supply and marketing differences.
Installation of plug-in electromagnetic flowmeter
The plug-in electromagnetic flowmeter sensor installation upstream straight pipe section is not less than 5D (D means pipe diameter), the downstream straight pipe section is not less than 2D:
1. Vertical installation. When the sensor is inserted into the pipeline, it should be less than 5° from the vertical diameter of the pipeline section. It is suitable for measuring the cleaning medium with small vibration of the pipeline.
2, inclined installation, the axis of the sensor and the axis of the pipe to be tested is 45 °, suitable for large pipe diameter and measuring the flow of liquid containing other impurities in the medium, the installation method is small, not suitable for entanglement.
There are two ways to insert the insertion point of the electromagnetic flowmeter sensor: one is inserted into the central axis of the pipe to be tested (ie 1/2 times the diameter of the pipe); the other is inserted into the inner wall of the pipe at 1/4 of the pipe.
1. Clean the welding slag and burrs of the base of the pipe to be tested.
2. Turn off the upstream flow control valve or use low pressure water supply.
3. Install the DN50 ball valve on the mounting base. Pay attention to the long cavity of the ball valve. Check whether the ball valve can be fully open and fully closed. Install the screw seat, compression nut and rubber sealing ring onto the ball valve and loosen the positioning nut. Insert the sensor into the rod through the ball valve and insert it into the pipe to be tested. At the same time, it should be noted that the direction of the sensor direction indicator rod should be consistent with the flow direction of the fluid. Ordering Information Please specify the name of the tested medium, working pressure, medium temperature, flow rate (flow rate) range, pipe diameter and the pipe network to be tested.