Because all kinds of ultrasonic flowmeters can be installed outside the pipe, non-contact flow measurement, the cost of the instrument is basically independent of the size of the pipe to be tested, while other types of flowmeters increase with the increase in caliber, so the cost is increased. The flowmeter is superior to the other functions of the same type of flowmeter. It is considered to be a good large-diameter flow measuring instrument. The Doppler ultrasonic flowmeter can measure the flow of two-phase medium, so it can be used for the measurement of dirty sewage such as sewers and sewage. In power plants, the use of portable ultrasonic flowmeters to measure large pipe diameters such as turbine water inflow and turbine circulating water is much more convenient than in the past. Ultrasonic flow juice can also be used for gas measurement. Pipe diameters range from 2cm to 5m, from a few meters wide open channels, culverts to 500m wide rivers.
In addition, the accuracy of the flow measurement of the ultrasonic measuring instrument is almost independent of the temperature, pressure, viscosity, density and other parameters of the measured fluid, and can be made into non-contact and portable measuring instruments, so it can solve the problem that other types of instruments are difficult to measure. Flow measurement problems for corrosive, non-conductive, radioactive, and flammable and explosive media. In addition, in view of the non-contact measurement characteristics, coupled with reasonable electronic circuits, one instrument can adapt to a variety of pipe diameter measurements and a variety of flow range measurements. The adaptability of ultrasonic flowmeters is also unmatched by other instruments. Ultrasonic flowmeters have some of the above advantages, so it has received more and more attention and has been developed into a series of products and generalization. It has been made into standard, high-temperature, explosion-proof and wet instruments of different channels to adapt to different media. Flow measurement for occasions and different pipeline conditions.
Installation of plug-in electromagnetic flowmeter
The plug-in electromagnetic flowmeter sensor installation upstream straight pipe section is not less than 5D (D means pipe diameter), the downstream straight pipe section is not less than 2D:
1. Vertical installation. When the sensor is inserted into the pipeline, it should be less than 5° from the vertical diameter of the pipeline section. It is suitable for measuring the cleaning medium with small vibration of the pipeline.
2, inclined installation, the axis of the sensor and the axis of the pipe to be tested is 45 °, suitable for large pipe diameter and measuring the flow of liquid containing other impurities in the medium, the installation method is small, not suitable for entanglement.
There are two ways to insert the insertion point of the electromagnetic flowmeter sensor: one is inserted into the central axis of the pipe to be tested (ie 1/2 times the diameter of the pipe); the other is inserted into the inner wall of the pipe at 1/4 of the pipe.
1. Clean the welding slag and burrs of the base of the pipe to be tested.
2. Turn off the upstream flow control valve or use low pressure water supply.
3. Install the DN50 ball valve on the mounting base. Pay attention to the long cavity of the ball valve. Check whether the ball valve can be fully open and fully closed. Install the screw seat, compression nut and rubber sealing ring onto the ball valve and loosen the positioning nut. Insert the sensor into the rod through the ball valve and insert it into the pipe to be tested. At the same time, it should be noted that the direction of the sensor direction indicator rod should be consistent with the flow direction of the fluid. Ordering Information Please specify the name of the tested medium, working pressure, medium temperature, flow rate (flow rate) range, pipe diameter and the pipe network to be tested.
Battery powered electromagnetic flowmeter features
1. Micro power consumption design, five 3.6V lithium batteries, continuous operation for 5~10 years, and easy to replace the battery.
2. The display adopts LCD large-screen liquid crystal display, which can display multiple flow parameters such as instantaneous flow rate, flow rate (pipe pressure), cumulative total amount of forward and reverse, and alarm prompt.
3. There are no movable and resistance parts in the measuring tube, no entanglement or clogging phenomenon, almost no pressure loss, and long-term reliable and continuous work.
4. Wide measurement range: The flow rate measurement range can reach 0.05m/s~10m/s.
5. High measurement accuracy: ±0.5% ~ ±1.0%.
6. Measurement stability: The measurement accuracy is not affected by changes in physical parameters such as temperature, pressure, viscosity and density of the measured medium.
7, with pulse output and GSM wireless data remote transmission.
8, with data storage function, up to 1000 groups of traffic data can be saved, and the data storage time can be arbitrarily set and data query.