﻿ Mechanical Water Flow Meter

# Mechanical Water Flow Meter

2020-05-17 06:39:05

## Mechanical Water Flow Meter

Instruments that measure fluid flow are collectively referred to as flow meters or flow meters. The flowmeter is one of the important instruments in industrial measurement. With the development of industrial production, the accuracy and range of flow measurement requirements are getting higher and higher, and the flow measurement technology is changing with each passing day. Various types of flow meters have been introduced to suit various applications. More than 100 flow meters have been put into use. From different perspectives, flow meters have different classification methods. There are two commonly used classification methods. One is to classify according to the measurement principle adopted by the flowmeter: the second is to classify according to the structural principle of the flowmeter.
Sort by measurement principle
a. Mechanical principle:
Instruments belonging to such principles have differential pressure type, rotor type using Bernoulli's theorem;
Impulse type using the momentum theorem, movable tube type;
Direct mass equation using Newton's second law;
a target using the principle of fluid momentum;
Turbine using the angular momentum theorem;
Vortex type using vortex principle of fluid oscillation;
Use the total static pressure difference of the pitot tube type as well as volumetric and sputum, trough and so on.
b. Electrical principle:
Electromagnetic,
Differential capacitor type,
Inductive,
Strain resistance type, etc.
c. Acoustic principle:
Ultrasonic.
d. Thermal principles:
Thermal type,
Direct calorimetry,
Indirect calorimetry and so on.

## Mechanical Water Flow Meter

However, since the volume of condensed water in the wet steam is relatively small, the flow of excess water in the ascending pipe often appears as an annular flow structure, but when the amount of water is particularly large, it also appears as a ring with fibers. Flow structure. Among them, the fibrous fluid is actually a condensed water. Flow in a vertically descending pipe: In a vertically descending pipe, the structure of the gas-liquid two-phase flow is similar to that of a vertically ascending flow, but differs not only in the opposite flow direction but also in the case of the same average flow velocity. The flow rate of the liquid in the vertical drop pipe is much faster than the flow rate of the liquid in the vertical rise pipe.
Second, steam vortex flowmeter measurement considerations
Vortex flowmeters measure steam. Accurately arrange the steam traps accurately: People have long discovered that when the steam has more water, the vortex flowmeter will have a "leakage pulse" phenomenon, that is, in the case of a stable steam flow rate, the vortex flow It should have a stable pulse output proportional to the flow rate. But sometimes it is found that the output pulse of the meter is inexplicably less. The distribution of the output pulse recorded on the two-dimensional coordinates can also be clearly seen. The pulse that should be approximately evenly distributed is one less at a place. Pulses, in severe cases, are a lot less pulses, and in the worst case, there are no pulses at all. This may be related to the formation of a vortex column by impinging on a large volume of droplets having a non-uniform distribution on the vortex generator.

## Mechanical Water Flow Meter

Intelligent GPRS MAG flow meter working principle
Since the GPRS communication is a data packet communication network based on an IP address, after the monitoring center computer logs in to the Internet network, the IP address is assigned by the Internet, and each GPRS monitoring point establishes a connection with the host through the address and communicates. Each monitoring point collects data through the data acquisition module, processes the data through the embedded processor in the GPRS terminal, encapsulates the protocol, sends it to the GPRS network, and transmits it to the user data monitoring center computer.
Battery-powered converter GPRS communication introduction
The battery-powered converter with GPRS function can directly send data to the remote control center through the GPRS wireless module GPRS-DY or GPRS-DF, hereinafter referred to as the GPRS module, and then store the collected data in the database at regular intervals. Through the system, even in an off-site away from the observation site, it is convenient to collect and read data of each monitoring point, and truly realize the functions of remote monitoring and data sharing.