Choice of caliber
The choice of the diameter of the electromagnetic flowmeter is not necessarily the same as the diameter of the pipe, and should be determined by the flow rate and flow rate. However, as the chemical raw materials and intermediate liquid of Salt Lake, the viscosity is large and the flow rate is low (generally 2.0 to 4.0 m/s). The electromagnetic flowmeter is used on such a pipe, and the diameter of the sensor is the same as the diameter of the pipe. If the flow rate is lower than 1.0m/s, an electromagnetic flowmeter of appropriate flow rate can be selected to ensure measurement accuracy and relatively reduce investment.
Selection of flow rate and range
Basically, they are anti-corrosion plastic pipes, and considering the lining of the flow meter, the flow rate is generally controlled at 2m/s. For some materials that are easy to crystallize (such as sodium carbonate solution, compounding liquid: crystallized mainly boron and magnesium double salt), the flow rate is increased to 3.0 to 4.0 m/s by taking measures. The flow rate will increase the flow noise, and the vibration of the pipeline will affect the measurement accuracy. Under the condition that the electromagnetic flowmeter is installed, the shock absorber should be installed before and after. The full scale of the meter is greater than the expected maximum flow value, which is typically 1.2 times the estimated maximum flow. The normal measurement flow is greater than 50% of the full scale of the meter to ensure a certain measurement accuracy.
Precautions before gas turbine flowmeter is enabled
Keywords: gas turbine flowmeter, Tianxin gas turbine flowmeter, gas gas turbine flowmeter
First, the precautions before the gas turbine flowmeter is enabled
In order to ensure trouble-free, safe start-up and commissioning of the ultrasonic flowmeter, the following pre-checks must be carried out:
1. Before starting the flowmeter officially, be sure to handle each leak point;
2. Check the power supply voltage on the terminal block and the corresponding positive and negative polarity;
3. Check the polarity of the frequency output lead connected to the RS485 converter;
4. Check whether the gas turbine flowmeter and temperature/pressure transmitter display are correct and reliable, including the installation direction and verticality of the flowmeter; the sealing performance of all joints;
5, wiring inspection: should be in accordance with the system wiring diagram provided by Jiangsu Tianxin Instrument Co., Ltd., check all wiring is correct.
Intelligent GPRS MAG flow meter working principle
Since the GPRS communication is a data packet communication network based on an IP address, after the monitoring center computer logs in to the Internet network, the IP address is assigned by the Internet, and each GPRS monitoring point establishes a connection with the host through the address and communicates. Each monitoring point collects data through the data acquisition module, processes the data through the embedded processor in the GPRS terminal, encapsulates the protocol, sends it to the GPRS network, and transmits it to the user data monitoring center computer.
Battery-powered converter GPRS communication introduction
The battery-powered converter with GPRS function can directly send data to the remote control center through the GPRS wireless module GPRS-DY or GPRS-DF, hereinafter referred to as the GPRS module, and then store the collected data in the database at regular intervals. Through the system, even in an off-site away from the observation site, it is convenient to collect and read data of each monitoring point, and truly realize the functions of remote monitoring and data sharing.