Choice of protection level
The protection level of the electromagnetic flowmeter should be selected according to the actual situation. If the sensor is required to be installed below the ground and is often flooded, IP68 should be selected. If the sensor is installed above the ground, IP65 and IP67 should be selected. In any case, the display area of ??the electromagnetic flowmeter cannot be used with water or enter the water, which may cause damage to the electromagnetic flowmeter. As a chemical company, the workshop produces high humidity, but IP65 (IP65 is a water-proof type. The electromagnetic flowmeter sensor housing allows the faucet to spray water from the sensor in any direction of the sensor of the electromagnetic flowmeter. The pressure of the water spray is 30Kpa, the water output It is 12.5L/S and the distance is 3 meters.) The electromagnetic flowmeter of the protection grade fully meets the requirements of on-site measurement.
Choice of connection method
Electromagnetic flowmeters have threaded connections, flanged connections, clamps, etc. In the chemical industry, flange-connected electromagnetic flowmeters are generally used. The selection must be consistent with the process-flange engineering pressure and standards.
Conclusion, with the maturity of the domestic instrument industry, chemical companies such as lithium carbonate, boric acid and potash will also enter the market in line with the domestic market. As an important industrial flow measuring instrument, electromagnetic flowmeter should play its role in DCS control. The right selection is especially important to provide accurate measurement data, reduce labor costs and increase production efficiency.
However, since the volume of condensed water in the wet steam is relatively small, the flow of excess water in the ascending pipe often appears as an annular flow structure, but when the amount of water is particularly large, it also appears as a ring with fibers. Flow structure. Among them, the fibrous fluid is actually a condensed water. Flow in a vertically descending pipe: In a vertically descending pipe, the structure of the gas-liquid two-phase flow is similar to that of a vertically ascending flow, but differs not only in the opposite flow direction but also in the case of the same average flow velocity. The flow rate of the liquid in the vertical drop pipe is much faster than the flow rate of the liquid in the vertical rise pipe.
Second, steam vortex flowmeter measurement considerations
Vortex flowmeters measure steam. Accurately arrange the steam traps accurately: People have long discovered that when the steam has more water, the vortex flowmeter will have a "leakage pulse" phenomenon, that is, in the case of a stable steam flow rate, the vortex flow It should have a stable pulse output proportional to the flow rate. But sometimes it is found that the output pulse of the meter is inexplicably less. The distribution of the output pulse recorded on the two-dimensional coordinates can also be clearly seen. The pulse that should be approximately evenly distributed is one less at a place. Pulses, in severe cases, are a lot less pulses, and in the worst case, there are no pulses at all. This may be related to the formation of a vortex column by impinging on a large volume of droplets having a non-uniform distribution on the vortex generator.
Selection of measurement types for special industrial and commercial users
At present, there are some special users in the supply of industrial and commercial users. On the one hand, the gas points are more dispersed, and the amount of gas used in a single combustion device is small (generally 2 to 9 m3/h); on the other hand, the total amount of gas used is Very large, there are many problems with the accurate measurement of the natural gas supply of such users. This type of industrial and commercial users have multiple fuel appliances with different gas consumption. Different gas consumption at different time periods brings certain difficulties to the accurate measurement of natural gas. Under the condition of satisfying the maximum gas consumption, it cannot be measured under the condition of small flow. The problem.
1. Case 1: A large shopping and entertainment mall
The mall has 22 gas points distributed on the second, fourth and fifth floors. In order to ensure the accuracy of natural gas measurement, the gas supply system has a master meter (turbine flow meter Q650) and 22 sets of compensation functions. As a metering table, the flow meter usually has a gas consumption of about 400m3/d and a holiday of about 800m3/d. It is found through the data collection that the total number of meters and sub-meters is about 17m3 per day; in the initial 54d. The cumulative gas consumption of the master meter and the minute meter differs by 11511.80 m3.