Choice of protection level
The protection level of the electromagnetic flowmeter should be selected according to the actual situation. If the sensor is required to be installed below the ground and is often flooded, IP68 should be selected. If the sensor is installed above the ground, IP65 and IP67 should be selected. In any case, the display area of ??the electromagnetic flowmeter cannot be used with water or enter the water, which may cause damage to the electromagnetic flowmeter. As a chemical company, the workshop produces high humidity, but IP65 (IP65 is a water-proof type. The electromagnetic flowmeter sensor housing allows the faucet to spray water from the sensor in any direction of the sensor of the electromagnetic flowmeter. The pressure of the water spray is 30Kpa, the water output It is 12.5L/S and the distance is 3 meters.) The electromagnetic flowmeter of the protection grade fully meets the requirements of on-site measurement.
Choice of connection method
Electromagnetic flowmeters have threaded connections, flanged connections, clamps, etc. In the chemical industry, flange-connected electromagnetic flowmeters are generally used. The selection must be consistent with the process-flange engineering pressure and standards.
Conclusion, with the maturity of the domestic instrument industry, chemical companies such as lithium carbonate, boric acid and potash will also enter the market in line with the domestic market. As an important industrial flow measuring instrument, electromagnetic flowmeter should play its role in DCS control. The right selection is especially important to provide accurate measurement data, reduce labor costs and increase production efficiency.
Precautions before gas turbine flowmeter is enabled
Keywords: gas turbine flowmeter, Tianxin gas turbine flowmeter, gas gas turbine flowmeter
First, the precautions before the gas turbine flowmeter is enabled
In order to ensure trouble-free, safe start-up and commissioning of the ultrasonic flowmeter, the following pre-checks must be carried out:
1. Before starting the flowmeter officially, be sure to handle each leak point;
2. Check the power supply voltage on the terminal block and the corresponding positive and negative polarity;
3. Check the polarity of the frequency output lead connected to the RS485 converter;
4. Check whether the gas turbine flowmeter and temperature/pressure transmitter display are correct and reliable, including the installation direction and verticality of the flowmeter; the sealing performance of all joints;
5, wiring inspection: should be in accordance with the system wiring diagram provided by Jiangsu Tianxin Instrument Co., Ltd., check all wiring is correct.
Intelligent GPRS MAG flow meter features
☆ Timing acquisition of electromagnetic flowmeter data: The electromagnetic flowmeter is used to send a data mode to the host every hour.
The data includes:
(1) Instrument status part: module address, module name, current module time, module battery status;
(2) Meter data part: instantaneous flow rate, instantaneous flow rate, fluid conductivity ratio, positive and negative cumulative value
☆ Support short message, fixed IP and dynamic IP address transmission. Data transmission is stable and reliable, low power consumption, and low battery alarm. SMS (SMS): Support all China Unicom and China Mobile SIM
card users; GPRS mode: support for the opening of GPRS wireless Internet access (GPRS access point must be CMNET) China Mobile SIM card users
☆ GPRS communication supports two modes of TCP and UDP, and supports two connection modes of IP address and dynamic domain name.
☆ Integration of acquisition and transmission, improving system reliability and reducing costs;
☆ Do not crash, automatically restore after dropped calls;
☆ GPRS remote online setting and short message remote setting function;
☆ Onboard industrial grade GSM/GPRS communication module, convenient for users to choose GSM/GPRS networking mode
☆ Provide user setting software, open interface
☆ Good waterproof performance