The lining material should be selected according to the corrosiveness, wear and temperature of the tested medium:
First, natural rubber (soft rubber)
1, better elasticity, wear resistance and tearing force
2, resistant to general weak acid, weak alkali corrosion
3. Water and sewage
Second, acid-resistant rubber (hard rubber)
It can withstand the corrosion of hydrochloric acid, acetic acid, oxalic acid, ammonia water, phosphoric acid and 50% sulfuric acid, sodium hydroxide and potassium hydroxide at normal temperature, but it is not resistant to corrosion by strong oxidants.
It can measure general acid, alkali and salt solutions.
Third, neoprene (Neoprene)
1, excellent elasticity, high tearing force, good wear resistance
2, resistant to general low concentration of acid and alkali, salt solution corrosion, but not resistant to oxidation of the medium, and temperature requirements <80 ° C;
It can measure water, sewage, mud and pulp.
Fourth, Polyurethane (Polyurethane)
1, excellent wear resistance
2, poor acid and alkali resistance, temperature requirements <40 ° C
Measure medium-strongly worn coal slurry, mud and slurry
Five, polytetrafluoroethylene (PTFE)
1. Boiling hydrochloric acid, sulfuric acid, nitric acid, aqua regia, concentrated alkali and various organic solvents
2, good wear resistance, poor bonding performance, temperature requirements -80 ~ +180 ° C;
Measuring concentration, concentrated alkali strong corrosive solution and sanitary medium
Gas turbine flowmeter and screwing vortex flowmeter measurement difference
Keywords: gas turbine flowmeter, Tianxin gas turbine flowmeter, gas gas turbine flowmeter
The gas turbine flowmeter has good repeatability and is the preferred flow meter for trade settlement. The gas swirling vortex flowmeter is generally repeatable.
The gas turbine flowmeter can obtain very high frequency signals with strong signal resolution. The frequency of the gas swirling vortex flowmeter is very low, and the smaller the diameter, the lower the signal resolution. The measurement error is to be measured.
The gas turbine flowmeter has a wide range, generally 40:1~20:1, and has a wide measuring range, which can be suitable for occasions with large flow changes. When natural gas is delivered, the flow changes are generally large and, therefore, are particularly suitable for natural gas measurements. The latter range is medium, generally 15:1 to 10:1, and the measurement range is slightly narrower. When measuring gas, it is more suitable for occasions where the flow rate does not change much.
In terms of seismic performance, gas turbine flowmeters have better resistance to pipeline vibration. The gas swirling vortex flowmeter is susceptible to pipeline vibration and should be taken care of when using it.
Intelligent GPRS MAG flow meter
The GPRS wireless transmission flowmeter is specially designed for the industry. It can meet the monitoring and management of traffic in urban water supply, heating and gas supply systems, and the processing and calculation of data. It is widely used in urban water supply, heating, gas supply systems, sewage treatment, Water conservancy projects and other industries.
GPRS wireless transmission flowmeter battery-powered converter is to use GPRS wireless communication to achieve remote monitoring. The data of the battery-powered converter can be read in any area covered by the GPRS wireless network. Through the GPRS wireless communication flowmeter network system, the data collected by the field instrument can be transmitted to the monitoring center in real time to achieve unified monitoring and distributed management of the field instrument.
GPRS wireless transmission flowmeter adopts specially designed sensor excitation system and high-performance lithium battery power supply system, and adopts 16-bit embedded ultra-micro power processor, full digital signal processing, stable measurement, high measurement accuracy and strong anti-interference ability. Features such as automatic bidirectional flow measurement, on-site instantaneous flow, positive and negativecumulative total display, self-diagnosis fault alarm, GSM data infinity transfer, etc. Users can realize traffic data through powerful GSM/GPRS remote management software system. GPRS wireless communication, storage, data analysis and other functions.