The lining material should be selected according to the corrosiveness, wear and temperature of the tested medium:
First, natural rubber (soft rubber)
1, better elasticity, wear resistance and tearing force
2, resistant to general weak acid, weak alkali corrosion
3. Water and sewage
Second, acid-resistant rubber (hard rubber)
It can withstand the corrosion of hydrochloric acid, acetic acid, oxalic acid, ammonia water, phosphoric acid and 50% sulfuric acid, sodium hydroxide and potassium hydroxide at normal temperature, but it is not resistant to corrosion by strong oxidants.
It can measure general acid, alkali and salt solutions.
Third, neoprene (Neoprene)
1, excellent elasticity, high tearing force, good wear resistance
2, resistant to general low concentration of acid and alkali, salt solution corrosion, but not resistant to oxidation of the medium, and temperature requirements <80 ° C;
It can measure water, sewage, mud and pulp.
Fourth, Polyurethane (Polyurethane)
1, excellent wear resistance
2, poor acid and alkali resistance, temperature requirements <40 ° C
Measure medium-strongly worn coal slurry, mud and slurry
Five, polytetrafluoroethylene (PTFE)
1. Boiling hydrochloric acid, sulfuric acid, nitric acid, aqua regia, concentrated alkali and various organic solvents
2, good wear resistance, poor bonding performance, temperature requirements -80 ~ +180 ° C;
Measuring concentration, concentrated alkali strong corrosive solution and sanitary medium
Liquid turbine flow meter classification and function description
Keywords: liquid turbine flowmeter, diesel liquid turbine flowmeter, pure water liquid turbine flowmeter
First, the liquid turbine flowmeter overview
The liquid turbine flowmeter is the main type of impeller flow (flow rate) meter, and the impeller type flowmeter also has an anemometer and a water meter. The TUF consists of a sensor and a conversion display. The sensor uses a multi-blade rotor to sense the average flow rate of the fluid, thereby deriving the flow or total amount. The rotational speed (or number of revolutions) of the rotor can be detected mechanically, magnetically, and photoelectrically and displayed and transmitted by the reading device.
Second, liquid turbine flowmeter classification
According to the classification of instrument functions, SDLWGY series turbine flowmeters can be divided into 2 categories.
1, turbine flow sensor / transmitter
2, intelligent integrated turbine flowmeter
Third, the liquid turbine flowmeter function description
1, turbine flow sensor / transmitter
This type of turbine flow product itself does not have a live display function, only the flow signal is transmitted far. The flow signal can be divided into pulse signal or current signal (4-20mA); the instrument is low in price, high in integration and small in size, and is especially suitable for use with computer control systems such as secondary display, PLC, DCS.
Intelligent GPRS MAG flow meter working principle
Since the GPRS communication is a data packet communication network based on an IP address, after the monitoring center computer logs in to the Internet network, the IP address is assigned by the Internet, and each GPRS monitoring point establishes a connection with the host through the address and communicates. Each monitoring point collects data through the data acquisition module, processes the data through the embedded processor in the GPRS terminal, encapsulates the protocol, sends it to the GPRS network, and transmits it to the user data monitoring center computer.
Battery-powered converter GPRS communication introduction
The battery-powered converter with GPRS function can directly send data to the remote control center through the GPRS wireless module GPRS-DY or GPRS-DF, hereinafter referred to as the GPRS module, and then store the collected data in the database at regular intervals. Through the system, even in an off-site away from the observation site, it is convenient to collect and read data of each monitoring point, and truly realize the functions of remote monitoring and data sharing.