1. Requirements for straight pipe sections
Inlet/outlet straight pipe section: inlet should be ≥10×DN; outlet should be ≥5×DN
2. Grounding point requirement
In order to make the instrument work reliably, improve the measurement accuracy, free from external parasitic potential / the sensor should have good grounding, the grounding resistance is less than 10. (If the metal pipe is well grounded, no special grounding device is required) 3.3 Pair installation Location requirements as shown
Insert the electromagnetic flowmeter according to the situation of the pipeline in the field. If the flowmeter without the ball valve is installed, it should be installed on the pipeline without overpressure (that is, the flowmeter without the ball valve can be installed without pressure), and the hole is opened in the pipeline. Diameter 50, ready to weld the connecting welded pipe to the opening of the pipe; for occasions requiring constant flow loading or unloading or no media spillage, a ball valve must be installed, that is, a plug-in electromagnetic flowmeter with a ball valve structure is selected; With a diameter of 50, it is ready to weld the connecting welded pipe to the opening of the pipe.
Measuring range : Recommended range: 0.5m/s to 10m/s continuously adjustable. Maximum use range: continuously adjustable from 0.2m/s to 15m/s
Signal output: 1, the switch quantity can be set to: pulse output (up to 1000HZ); high/low flow alarm; empty pipe alarm; flow direction indication;
Fault alarm; 2, current output: 4-20mA output
Configuration method: 1. Field configuration through three manual keys. 2. Field configuration via remote control. 3. Perform on-site configuration through the handheld communicator.
Memory: The memory that the EEPROM does not disappear, no battery saving.
Fourth, the liquid turbine flowmeter put into operation the opening and closing sequence
1. The sequence of opening and closing of the operation
For flow sensors without a bypass line, first open the flow sensor upstream valve at a medium opening and then slowly open the downstream valve. Run at a small flow rate for a period of time (eg 10 minutes), then fully open the upstream valve, then open the large downstream valve opening to adjust to the required normal flow.
The flow sensor equipped with the bypass pipe first opens the bypass pipe valve to open the upstream valve at a medium opening degree, slowly opens the downstream valve, and closes the opening of the small bypass valve to make the instrument run for a period of time with a small flow rate. Then fully open the upstream valve, fully close the bypass valve (to ensure no leakage), and finally adjust the downstream valve opening to the required flow.
2. Activation of low temperature and high temperature fluids
The low-temperature fluid pipeline should drain the water in the pipeline before the flow, and then run for 15 minutes at a small flow rate, and then gradually increase to the normal flow. Slow down when stopping, so that the pipe temperature and ambient temperature are gradually approaching. High temperature fluid operation is similar to this.
Intelligent GPRS MAG flow meter features
☆ Timing acquisition of electromagnetic flowmeter data: The electromagnetic flowmeter is used to send a data mode to the host every hour.
The data includes:
(1) Instrument status part: module address, module name, current module time, module battery status;
(2) Meter data part: instantaneous flow rate, instantaneous flow rate, fluid conductivity ratio, positive and negative cumulative value
☆ Support short message, fixed IP and dynamic IP address transmission. Data transmission is stable and reliable, low power consumption, and low battery alarm. SMS (SMS): Support all China Unicom and China Mobile SIM
card users; GPRS mode: support for the opening of GPRS wireless Internet access (GPRS access point must be CMNET) China Mobile SIM card users
☆ GPRS communication supports two modes of TCP and UDP, and supports two connection modes of IP address and dynamic domain name.
☆ Integration of acquisition and transmission, improving system reliability and reducing costs;
☆ Do not crash, automatically restore after dropped calls;
☆ GPRS remote online setting and short message remote setting function;
☆ Onboard industrial grade GSM/GPRS communication module, convenient for users to choose GSM/GPRS networking mode
☆ Provide user setting software, open interface
☆ Good waterproof performance