Accuracy class and function According to the measurement requirements and the use occasions, the instrument accuracy level is selected to achieve economical efficiency. For example, in the case of trade settlement, product handover and energy measurement, the accuracy level should be higher, such as 1.0, 0.5, or higher; for process control, select different accuracy levels according to control requirements; It is to detect the process flow, no need to do precise control and measurement, you can choose a lower accuracy level, such as 1.5, 2.5, or even 4.0, then you can use a low-cost plug-in electromagnetic flowmeter.
Measuring medium flow rate, meter range and diameter When measuring general medium, the full flow of electromagnetic flowmeter can be selected within the range of 0.5-12m/s of measuring medium flow, and the range is wider. The meter specification (caliber) is not necessarily the same as the process pipeline. It should be determined whether the measured flow range is within the flow rate range. That is, when the pipeline flow rate is too low to meet the flow meter requirements or the measurement accuracy cannot be guaranteed at this flow rate, It is necessary to reduce the gauge diameter, thereby increasing the flow rate inside the tube and obtaining satisfactory measurement results.
Try to avoid ferromagnetic objects and equipment with strong electromagnetic fields to prevent the magnetic field from affecting the working magnetic field and flow signal of the sensor.
Should be installed in the dry and ventilated place, to avoid sun and rain, the ambient temperature should be -20 ~ +60 ° C, relative humidity is less than 85%.
There should be ample space around the flowmeter for easy testing and maintenance.
According to different output signals, these products can be divided into LWGY-□N type and LWGY-□A type.
LWGY-□N type sensor: 12～24VDC power supply, three-wire pulse output, high level ≥8V, low level ≤0.8V; signal transmission distance ≤1000m;
LWGY-□A type transmitter: 24VDC power supply, 2-wire 4-20mA output, signal transmission distance ≤1000.
This type of turbine flow products are divided into basic type and explosion-proof type (ExdIIBT6).
2, intelligent integrated turbine flowmeter
Intelligent integrated turbine flowmeter adopts advanced ultra-low-power single-chip microcomputer technology to develop a new intelligent instrument integrating turbine flow sensor and display integration. It adopts double-row liquid crystal field display, with compact structure, intuitive reading and reliability. High, free from external power interference, lightning strike, low cost and other obvious advantages. The instrument has three-point correction of the meter coefficient, the intelligent compensation meter coefficient is nonlinear, and field correction can be performed. The high-definition LCD displays both instantaneous flow (4 significant digits) and cumulative flow (8 significant digits with clear function). All valid data is not lost after 10 years of power failure. These turbine flowmeters are explosion-proof products with an explosion-proof rating of ExdIIBT6.
Intelligent GPRS MAG flow meter working principle
Since the GPRS communication is a data packet communication network based on an IP address, after the monitoring center computer logs in to the Internet network, the IP address is assigned by the Internet, and each GPRS monitoring point establishes a connection with the host through the address and communicates. Each monitoring point collects data through the data acquisition module, processes the data through the embedded processor in the GPRS terminal, encapsulates the protocol, sends it to the GPRS network, and transmits it to the user data monitoring center computer.
Battery-powered converter GPRS communication introduction
The battery-powered converter with GPRS function can directly send data to the remote control center through the GPRS wireless module GPRS-DY or GPRS-DF, hereinafter referred to as the GPRS module, and then store the collected data in the database at regular intervals. Through the system, even in an off-site away from the observation site, it is convenient to collect and read data of each monitoring point, and truly realize the functions of remote monitoring and data sharing.