Accuracy class and function According to the measurement requirements and the use occasions, the instrument accuracy level is selected to achieve economical efficiency. For example, in the case of trade settlement, product handover and energy measurement, the accuracy level should be higher, such as 1.0, 0.5, or higher; for process control, select different accuracy levels according to control requirements; It is to detect the process flow, no need to do precise control and measurement, you can choose a lower accuracy level, such as 1.5, 2.5, or even 4.0, then you can use a low-cost plug-in electromagnetic flowmeter.
Measuring medium flow rate, meter range and diameter When measuring general medium, the full flow of electromagnetic flowmeter can be selected within the range of 0.5-12m/s of measuring medium flow, and the range is wider. The meter specification (caliber) is not necessarily the same as the process pipeline. It should be determined whether the measured flow range is within the flow rate range. That is, when the pipeline flow rate is too low to meet the flow meter requirements or the measurement accuracy cannot be guaranteed at this flow rate, It is necessary to reduce the gauge diameter, thereby increasing the flow rate inside the tube and obtaining satisfactory measurement results.
Try to avoid ferromagnetic objects and equipment with strong electromagnetic fields to prevent the magnetic field from affecting the working magnetic field and flow signal of the sensor.
Should be installed in the dry and ventilated place, to avoid sun and rain, the ambient temperature should be -20 ~ +60 ° C, relative humidity is less than 85%.
There should be ample space around the flowmeter for easy testing and maintenance.
Fourth, the liquid turbine flowmeter put into operation the opening and closing sequence
1. The sequence of opening and closing of the operation
For flow sensors without a bypass line, first open the flow sensor upstream valve at a medium opening and then slowly open the downstream valve. Run at a small flow rate for a period of time (eg 10 minutes), then fully open the upstream valve, then open the large downstream valve opening to adjust to the required normal flow.
The flow sensor equipped with the bypass pipe first opens the bypass pipe valve to open the upstream valve at a medium opening degree, slowly opens the downstream valve, and closes the opening of the small bypass valve to make the instrument run for a period of time with a small flow rate. Then fully open the upstream valve, fully close the bypass valve (to ensure no leakage), and finally adjust the downstream valve opening to the required flow.
2. Activation of low temperature and high temperature fluids
The low-temperature fluid pipeline should drain the water in the pipeline before the flow, and then run for 15 minutes at a small flow rate, and then gradually increase to the normal flow. Slow down when stopping, so that the pipe temperature and ambient temperature are gradually approaching. High temperature fluid operation is similar to this.
Intelligent GPRS MAG flow meter
The GPRS wireless transmission flowmeter is specially designed for the industry. It can meet the monitoring and management of traffic in urban water supply, heating and gas supply systems, and the processing and calculation of data. It is widely used in urban water supply, heating, gas supply systems, sewage treatment, Water conservancy projects and other industries.
GPRS wireless transmission flowmeter battery-powered converter is to use GPRS wireless communication to achieve remote monitoring. The data of the battery-powered converter can be read in any area covered by the GPRS wireless network. Through the GPRS wireless communication flowmeter network system, the data collected by the field instrument can be transmitted to the monitoring center in real time to achieve unified monitoring and distributed management of the field instrument.
GPRS wireless transmission flowmeter adopts specially designed sensor excitation system and high-performance lithium battery power supply system, and adopts 16-bit embedded ultra-micro power processor, full digital signal processing, stable measurement, high measurement accuracy and strong anti-interference ability. Features such as automatic bidirectional flow measurement, on-site instantaneous flow, positive and negativecumulative total display, self-diagnosis fault alarm, GSM data infinity transfer, etc. Users can realize traffic data through powerful GSM/GPRS remote management software system. GPRS wireless communication, storage, data analysis and other functions.