Ultrasonic flowmeter classification
Plug-in ultrasonic flowmeter: can be installed and maintained without interruption. Ceramic sensors are used for non-stop production installation using dedicated drilling equipment. Generally for mono measurement, in order to improve measurement accuracy, three channels can be selected.
Pipe-type ultrasonic flowmeter: the pipeline installation needs to be cut, but the subsequent maintenance can be stopped. Mono or 3-channel sensors are available.
External clip-on ultrasonic flowmeter: capable of performing fixed and mobile measurements. It is installed with a special coupling agent (silicone rubber cured at room temperature or high-temperature long-chain polymer grease) and does not damage the pipeline during installation.
Portable Ultrasonic Flowmeter: Portable, built-in rechargeable lithium battery, suitable for mobile measurement, with magnetic sensor.
1, non-contact measurement method, small size, easy to carry
2, suitable for on-site measurement of various sizes of pipe sound guiding media
3, built-in nickel-metal hydride rechargeable battery working time of more than 20 hours
4, user interface is flexible, easy to use
5, intelligent on-site printing function to ensure the integrity of the flow data
6, equipped with an integrated aluminum alloy protective box, can be used in harsh outdoor environments
Hand-held ultrasonic flowmeter: small size, light weight, built-in rechargeable lithium battery, hand-held, with magnetic sensor.
Explosion-proof ultrasonic flowmeter: used for explosive liquid flow measurement, it is explosion-proof and intrinsically safe. That is, the converter is explosion-proof and the sensor is intrinsically safe.
Turbine flowmeter composition
Turbine flowmeters are speed flowmeters, also known as impeller flowmeters. The impeller type flowmeter utilizes the relationship between the rotational angular velocity of the impeller placed in the fluid and the fluid flow velocity. By measuring the rotational speed of the impeller to reflect the volumetric flow rate of the fluid passing through the pipeline, it is a relatively mature high-accuracy instrument in the flowmeter. One. Turbine flowmeters generally consist of the following five typical parts:
The material of the body is generally steel or cast iron, and the ends are flanged. The small diameter gauge also has a threaded interface.
2. Measured component
The turbine has precision machined blades that together with a set of reduction gears and bearings form the measuring assembly. The two high-precision stainless steel permanent self-lubricating bearings that support the turbine ensure a long service life of the assembly. Turbine flowmeters can also be lubricated with an external lubricant pump, but be careful not to overdo it.
The following important information is available on the counter panel:
(1) Maximum working temperature and pressure:
(2) Metering and minimum and maximum flow levels
(3) Product model and number
(4) Explosion-proof grades and signs
(5) The equivalent of the fluid corresponding to the low frequency or high frequency pulse and the wiring method.
The rectifier is used to keep the fluid flowing through the turbine flow meter in a regular state, thereby eliminating the disturbance and having an unaffected effect on the metering. High metering accuracy.
5. Magnetic coupling transmission device
The device separates the counter portion in the atmospheric environment from the gas being measured and transmits the rotation of the measuring assembly to the counter.
Selection of measurement types for special industrial and commercial users
At present, there are some special users in the supply of industrial and commercial users. On the one hand, the gas points are more dispersed, and the amount of gas used in a single combustion device is small (generally 2 to 9 m3/h); on the other hand, the total amount of gas used is Very large, there are many problems with the accurate measurement of the natural gas supply of such users. This type of industrial and commercial users have multiple fuel appliances with different gas consumption. Different gas consumption at different time periods brings certain difficulties to the accurate measurement of natural gas. Under the condition of satisfying the maximum gas consumption, it cannot be measured under the condition of small flow. The problem.
1. Case 1: A large shopping and entertainment mall
The mall has 22 gas points distributed on the second, fourth and fifth floors. In order to ensure the accuracy of natural gas measurement, the gas supply system has a master meter (turbine flow meter Q650) and 22 sets of compensation functions. As a metering table, the flow meter usually has a gas consumption of about 400m3/d and a holiday of about 800m3/d. It is found through the data collection that the total number of meters and sub-meters is about 17m3 per day; in the initial 54d. The cumulative gas consumption of the master meter and the minute meter differs by 11511.80 m3.