Ultrasonic flowmeter classification
Plug-in ultrasonic flowmeter: can be installed and maintained without interruption. Ceramic sensors are used for non-stop production installation using dedicated drilling equipment. Generally for mono measurement, in order to improve measurement accuracy, three channels can be selected.
Pipe-type ultrasonic flowmeter: the pipeline installation needs to be cut, but the subsequent maintenance can be stopped. Mono or 3-channel sensors are available.
External clip-on ultrasonic flowmeter: capable of performing fixed and mobile measurements. It is installed with a special coupling agent (silicone rubber cured at room temperature or high-temperature long-chain polymer grease) and does not damage the pipeline during installation.
Portable Ultrasonic Flowmeter: Portable, built-in rechargeable lithium battery, suitable for mobile measurement, with magnetic sensor.
1, non-contact measurement method, small size, easy to carry
2, suitable for on-site measurement of various sizes of pipe sound guiding media
3, built-in nickel-metal hydride rechargeable battery working time of more than 20 hours
4, user interface is flexible, easy to use
5, intelligent on-site printing function to ensure the integrity of the flow data
6, equipped with an integrated aluminum alloy protective box, can be used in harsh outdoor environments
Hand-held ultrasonic flowmeter: small size, light weight, built-in rechargeable lithium battery, hand-held, with magnetic sensor.
Explosion-proof ultrasonic flowmeter: used for explosive liquid flow measurement, it is explosion-proof and intrinsically safe. That is, the converter is explosion-proof and the sensor is intrinsically safe.
Flowmeter product features:
Using intelligent judgment, the measurement correction setting is not used, and the air traffic control alarm and electrode detection application are more convenient;
Advanced "rough error handling" technology, which can remove fluids such as slurry to measure sharp disturbances, reduce output runout, maintain high precision measurement and make output more stable;
With a fluid density setting, it can display mass flow;
Constant current excitation current range, 125mA, 250mA optional, can be used with different manufacturers, different types of electromagnetic flow sensors;
Control function with remote reset of the totalizer, with contact signal input for starting and stopping accumulation, suitable for total inspection and batch processing applications;
With self-test and self-diagnosis function;
Advanced non-volatile memory for higher circuit reliability and effective protection of setup and measurement parameters;
The meter can be equipped with an unpowered clock and memory for recording the power down time, power-on time and power-down time;
The meter has an optional hour recording function that can store flow and electrode resistance measurements for more than 30 days.
The new keyboard processing method avoids the keyboard operation affecting the measurement, and can enter and return the operation menu to make the parameter setting more convenient;
The total display uses 10-bit decimal 9999999999 full-value carry, which solves the practice of double-word full value 4294967285 (hexadecimal FFFF) carry-in habit;
With wireless transmission, the mesh network is organized by the wireless HART protocol.
Selection of measurement types for special industrial and commercial users
2. Case 2: A heavy machine tool factory
The main gas plant of the heavy-duty machine tool factory is 13 production kiln, and there is also a machine repair workshop and 63 mobile burners. The total gas load of all gas facilities is 1837m3/h. See Table 1 for details. The customer promised that all gas facilities could not be fully activated at the same time, the maximum gas consumption required was 1200m3/h, and the minimum pressure required before the furnace was 0.055MPa. The pressure regulator used in the on-site pressure regulating metering box is opened and ready, and the flow meter is in the form of a G1600 turbine table plus bypass.
When debugging the kiln, it was found that the flowmeter in the pressure regulating metering box had a stop-and-go phenomenon. If only some of the kiln was opened or only some of the kiln was used, there would be a phenomenon that the flowmeter did not measure or the measurement error was large.