Ultrasonic flowmeter classification
Plug-in ultrasonic flowmeter: can be installed and maintained without interruption. Ceramic sensors are used for non-stop production installation using dedicated drilling equipment. Generally for mono measurement, in order to improve measurement accuracy, three channels can be selected.
Pipe-type ultrasonic flowmeter: the pipeline installation needs to be cut, but the subsequent maintenance can be stopped. Mono or 3-channel sensors are available.
External clip-on ultrasonic flowmeter: capable of performing fixed and mobile measurements. It is installed with a special coupling agent (silicone rubber cured at room temperature or high-temperature long-chain polymer grease) and does not damage the pipeline during installation.
Portable Ultrasonic Flowmeter: Portable, built-in rechargeable lithium battery, suitable for mobile measurement, with magnetic sensor.
1, non-contact measurement method, small size, easy to carry
2, suitable for on-site measurement of various sizes of pipe sound guiding media
3, built-in nickel-metal hydride rechargeable battery working time of more than 20 hours
4, user interface is flexible, easy to use
5, intelligent on-site printing function to ensure the integrity of the flow data
6, equipped with an integrated aluminum alloy protective box, can be used in harsh outdoor environments
Hand-held ultrasonic flowmeter: small size, light weight, built-in rechargeable lithium battery, hand-held, with magnetic sensor.
Explosion-proof ultrasonic flowmeter: used for explosive liquid flow measurement, it is explosion-proof and intrinsically safe. That is, the converter is explosion-proof and the sensor is intrinsically safe.
Working principle and sensor parameters of ultrasonic external clamp flow meter
External clip type ultrasonic flow meter external clip type ultrasonic flow meter series measuring full flow of liquid flow meter, optional explosion-proof and intrinsically safe type, explosion-proof grade Exd[ib]IIBT6, and optional intrinsically safe remote control, explosion-proof grade ExibIIBT6, measurable medium: drinking water, river water, sea water, ground water cooling water, high temperature water, sewage, lubricating oil, diesel, fuel chemical liquid, other homogeneous fluid.
External clip sensor parameters:
The external clip type sensor is an installation method in which the sensor is directly bundled on the outer surface of the pipe to be measured to realize flow measurement, and has a diameter independent of the pipe diameter.
Simple installation, no need to stop production, no pressure loss and so on.
Intelligent GPRS MAG flow meter working principle
Since the GPRS communication is a data packet communication network based on an IP address, after the monitoring center computer logs in to the Internet network, the IP address is assigned by the Internet, and each GPRS monitoring point establishes a connection with the host through the address and communicates. Each monitoring point collects data through the data acquisition module, processes the data through the embedded processor in the GPRS terminal, encapsulates the protocol, sends it to the GPRS network, and transmits it to the user data monitoring center computer.
Battery-powered converter GPRS communication introduction
The battery-powered converter with GPRS function can directly send data to the remote control center through the GPRS wireless module GPRS-DY or GPRS-DF, hereinafter referred to as the GPRS module, and then store the collected data in the database at regular intervals. Through the system, even in an off-site away from the observation site, it is convenient to collect and read data of each monitoring point, and truly realize the functions of remote monitoring and data sharing.