Daily maintenance (1)
Only need to periodically check the instrument, check the environment around the instrument, remove the dirt, ensure that no water and other substances are inspected, check whether the wiring is good, check whether there is a new strong electromagnetic field device near the instrument or a new installed wire across the instrument. If the measuring medium is easy to contaminate the electrode or precipitate or scale in the measuring tube wall, it should be regularly cleaned and cleaned.
Folding fault finding
After the flowmeter starts to be put into operation or is put into operation for a period of time, it is found that the instrument is not working properly. First, check the external condition of the flowmeter, such as whether the power supply is good, whether the pipeline is leaking or is in a state of non-full pipe, whether there is air bubble in the pipe, whether the signal cable is Damage, converter output signal (ie rear position meter input loop) is open circuit. Remember to blindly disassemble the flowmeter.
Advantages of turbine flow meters
The pressure loss is small, and the impeller has anti-corrosion function;
1.Adopt advanced ultra-low power single-chip microcomputer technology, the whole machine has strong functions, low power consumption and superior performance.
2.Intelligent flow display with nonlinear accuracy compensation. Correction formula accuracy is better than ±0.02%
3.The meter factor can be set online by button and can be displayed on the LCD screen. The LCD screen is intuitive and clear, and the reliability is strong.
4.EEPROM is used to accumulate the flow and the meter coefficient is powered down. The protection time is more than 10 years.
5.Using high-performance MCU central processing unit to complete data acquisition and processing display output, cumulative flow instantaneous flow
6.The screen displays a convenient man-machine interface for data transfer in standard 485 format.
7.Fully hard alloy (tungsten carbide) shielded cantilever beam structure bearing, integrating rotary bearing and pressure bearing, greatly improving bearing life and working in medium with a small amount of sediment and dirt.
8.It adopts 1Cr18Ni9Ti all stainless steel structure (the turbine adopts 2Cr13) and has good anti-corrosion performance.
9.Easy to maintain, self-rectifying structure, small and light, simple structure, which can be combined in a short time.
10.Internal cleaning is simple.
11.Strong anti-magnetic interference and vibration capability, reliable performance and long life
12.The lower limit flow rate is low and the measuring range is wide.
13.The on-site display type LCD screen is clear and intuitive, with low power consumption. The 3V lithium battery can be operated continuously for more than 5 years.
Corrosion resistant, suitable for acid and alkali solutions
Selection of measurement types for special industrial and commercial users
2. Case 2: A heavy machine tool factory
The main gas plant of the heavy-duty machine tool factory is 13 production kiln, and there is also a machine repair workshop and 63 mobile burners. The total gas load of all gas facilities is 1837m3/h. See Table 1 for details. The customer promised that all gas facilities could not be fully activated at the same time, the maximum gas consumption required was 1200m3/h, and the minimum pressure required before the furnace was 0.055MPa. The pressure regulator used in the on-site pressure regulating metering box is opened and ready, and the flow meter is in the form of a G1600 turbine table plus bypass.
When debugging the kiln, it was found that the flowmeter in the pressure regulating metering box had a stop-and-go phenomenon. If only some of the kiln was opened or only some of the kiln was used, there would be a phenomenon that the flowmeter did not measure or the measurement error was large.