Daily maintenance (1)
Only need to periodically check the instrument, check the environment around the instrument, remove the dirt, ensure that no water and other substances are inspected, check whether the wiring is good, check whether there is a new strong electromagnetic field device near the instrument or a new installed wire across the instrument. If the measuring medium is easy to contaminate the electrode or precipitate or scale in the measuring tube wall, it should be regularly cleaned and cleaned.
Folding fault finding
After the flowmeter starts to be put into operation or is put into operation for a period of time, it is found that the instrument is not working properly. First, check the external condition of the flowmeter, such as whether the power supply is good, whether the pipeline is leaking or is in a state of non-full pipe, whether there is air bubble in the pipe, whether the signal cable is Damage, converter output signal (ie rear position meter input loop) is open circuit. Remember to blindly disassemble the flowmeter.
Flowmeter product features:
Using intelligent judgment, the measurement correction setting is not used, and the air traffic control alarm and electrode detection application are more convenient;
Advanced "rough error handling" technology, which can remove fluids such as slurry to measure sharp disturbances, reduce output runout, maintain high precision measurement and make output more stable;
With a fluid density setting, it can display mass flow;
Constant current excitation current range, 125mA, 250mA optional, can be used with different manufacturers, different types of electromagnetic flow sensors;
Control function with remote reset of the totalizer, with contact signal input for starting and stopping accumulation, suitable for total inspection and batch processing applications;
With self-test and self-diagnosis function;
Advanced non-volatile memory for higher circuit reliability and effective protection of setup and measurement parameters;
The meter can be equipped with an unpowered clock and memory for recording the power down time, power-on time and power-down time;
The meter has an optional hour recording function that can store flow and electrode resistance measurements for more than 30 days.
The new keyboard processing method avoids the keyboard operation affecting the measurement, and can enter and return the operation menu to make the parameter setting more convenient;
The total display uses 10-bit decimal 9999999999 full-value carry, which solves the practice of double-word full value 4294967285 (hexadecimal FFFF) carry-in habit;
With wireless transmission, the mesh network is organized by the wireless HART protocol.
Second, the gas turbine flowmeter projection operation steps
1. Open the bypass shut-off valve;
2. Open the flow upstream shutoff valve;
3. Slowly open the downstream shutoff valve of the flowmeter;
4. Slowly close the bypass shutoff valve.
Third, the gas turbine flowmeter stop table operation steps
1. Open the bypass shut-off valve;
2. Turn off the downstream shutoff valve of the flowmeter;
3. Close the flow upstream shutoff valve;
Fourth, gas turbine flowmeter use precautions
1. The newly installed or repaired pipeline must be purged. When purging the metering line, the flow meter must be removed and the corresponding short section replaced the flow meter for purging.
2. When the gas turbine flowmeter pipeline is put into production, it should be slowly boosted and gradually increase the flow rate. When the production is stopped, it should be slowly depressurized.
3. Check the sound of the gas turbine flowmeter or the vibration of the casing during operation to determine whether the turbine blades and bearings are working properly. At low flow rates, attention should be paid to the change in sound, and the vibration of the casing is observed at high flow rates.