Daily maintenance (1)
Only need to periodically check the instrument, check the environment around the instrument, remove the dirt, ensure that no water and other substances are inspected, check whether the wiring is good, check whether there is a new strong electromagnetic field device near the instrument or a new installed wire across the instrument. If the measuring medium is easy to contaminate the electrode or precipitate or scale in the measuring tube wall, it should be regularly cleaned and cleaned.
Folding fault finding
After the flowmeter starts to be put into operation or is put into operation for a period of time, it is found that the instrument is not working properly. First, check the external condition of the flowmeter, such as whether the power supply is good, whether the pipeline is leaking or is in a state of non-full pipe, whether there is air bubble in the pipe, whether the signal cable is Damage, converter output signal (ie rear position meter input loop) is open circuit. Remember to blindly disassemble the flowmeter.
Ultrasonic flowmeter performance characteristics
Measuring liquid: single fluid such as water, seawater, alkali, etc., turbidity less than 10000ppm, particle size less than 1mm
Pipe material: Ultrasonic penetrating full pipe such as steel, cast iron, PVC pipe, dense texture, no serious corrosion and scaling, allowing thin layer of dense lining
Pipe lining material: rubber, epoxy asphalt, fiberglass, mortar, etc., or unlined
Pipe diameter range: DN15mm~6000mm, plug-in sensor for pipe diameter DN25mm~6000mm
Flow rate range: -32m/s~0~+32m/s, bidirectional flow
Accuracy: ±1.0% of displayed value (under standard conditions). Configure plug-in sensor accuracy up to 0.5%
Signal output: 4-20mA or 0-20mA RS-485
Sensor: It is divided into external clip type and plug-in type. The standard configuration is clip-on. The maximum operating pressure of the plug-in sensor is 3 MPa.
Straight pipe length: upstream >10D, downstream >5D from pump outlet >30D (D: pipe diameter)
Display mode: 2*20 character LCD display with LED backlight
Protection level: IP65
Explosion-proof grade: ExdIIBT4
Fixed ultrasonic flowmeter power supply: AC110V, AC220V, DC8~36V or AC7~30V
Ambient temperature / humidity: Host -20 ° C ~ +80 ° C; sensor -20 ° C ~ +80 ° C or -20 ° C ~ +160 ° C optional, the sensor can be immersed in water, water depth less than 2m.
Intelligent GPRS MAG flow meter working principle
Since the GPRS communication is a data packet communication network based on an IP address, after the monitoring center computer logs in to the Internet network, the IP address is assigned by the Internet, and each GPRS monitoring point establishes a connection with the host through the address and communicates. Each monitoring point collects data through the data acquisition module, processes the data through the embedded processor in the GPRS terminal, encapsulates the protocol, sends it to the GPRS network, and transmits it to the user data monitoring center computer.
Battery-powered converter GPRS communication introduction
The battery-powered converter with GPRS function can directly send data to the remote control center through the GPRS wireless module GPRS-DY or GPRS-DF, hereinafter referred to as the GPRS module, and then store the collected data in the database at regular intervals. Through the system, even in an off-site away from the observation site, it is convenient to collect and read data of each monitoring point, and truly realize the functions of remote monitoring and data sharing.