Main features of vortex flowmeter
Measuring medium: nominal diameter of liquid, gas and steam: DN15-DN300 (non-standard products can be customized according to user requirements) Temperature range: -40°C~350°C Pressure specification: PN1.6Mpa; PN2.5Mpa; PN4.0Mpa, Higher pressure specifications can be customized to a specific range: normal range 1:10 Extended range 1:15 Pressure loss factor: Cd ≤ 2.6 System measurement accuracy: liquid, gas indication ± 1%, steam indication ± 1.5% plug-in flow The measured value is ±2.5% of the supply voltage: sensor +12VDC, +24VDC (optional) transmitter +24VDC.
On-site display type The meter comes with 3.6 lithium battery output signal: sensor pulse frequency signal 0.1~3000Hz low level ≤1V high level ≥6V.
Transmitter two-wire 4~20mADC current signal for vibration acceleration: Piezoelectric ≤0.2g Ambient temperature: -40°C~55°C (non-explosion-proof place) -20°C~55°C (explosion-proof place) Ambient humidity: Relative Humidity 5~85% signal remote transmission distance: ≤500m signal line interface: internal thread M20×1.5 explosion-proof grade: iaIICT2-T5 protection grade: ordinary IP65 submersible IP68 instrument material: converter shell is made of aluminum alloy, the body part is adopted 1Cr18Ni9Ti can also be made of special materials according to user requirements.
The main technical parameters of ultrasonic external clamp flowmeter:
Measurement accuracy: better than 1%
Repeatability: better than 0.2%
Measurement period: 500ms (2 times per second, 128 sets of data collected per cycle)
Working power: 220VAC/8~36VDC
Maximum flow rate: 64m / s (flow rate resolution 0.001m / s)
Display: 2 × 10 Chinese characters backlit LCD can display instantaneous flow and positive, negative, net cumulative flow, flow rate, etc.
Operation: 4×4 touch keyboard (F4 main unit magnetic 4 button) operation
Signal input: 3 channels of 4-20mA analog input, accuracy 0.1%, can input pressure, liquid level, temperature and other signals
2-way three-wire PT100 platinum resistor
Signal output: 1 isolated RS485 output
1 way 4-20mA or 0-20mA output
1 channel isolated OCT (programmable between pulse width 6~1000ms, default 200ms)
1 relay output (pulse width 200ms)
Data storage: Optional built-in data storage (SD card) can store time, instantaneous flow, accumulated flow, signal status, etc., through special software, can import data into computer for statistical and management
Communication protocol: MODBUS protocol, M-BUS protocol, FUJI extension protocol, and compatible with communication protocols of similar products of other domestic manufacturers
Other functions: automatic memory before 512 days, the first 128 months, the first 10 years of positive / negative / net cumulative flow
Automatically remember the first 30 times of power-on and power-off time and flow rate, and can realize automatic or manual addition of traffic, which can be read by MODBUS protocol.
Programmable batch (quantitative) controller, fault self-diagnosis function
Software upgrade via code file transmitted via E-mail
Protection level: sensor IP68, F4 host IP68, remaining host IP65
Explosion-proof grade: EXdIIBT4 (JN-100F2 type)
Installation of plug-in electromagnetic flowmeter
The plug-in electromagnetic flowmeter sensor installation upstream straight pipe section is not less than 5D (D means pipe diameter), the downstream straight pipe section is not less than 2D:
1. Vertical installation. When the sensor is inserted into the pipeline, it should be less than 5° from the vertical diameter of the pipeline section. It is suitable for measuring the cleaning medium with small vibration of the pipeline.
2, inclined installation, the axis of the sensor and the axis of the pipe to be tested is 45 °, suitable for large pipe diameter and measuring the flow of liquid containing other impurities in the medium, the installation method is small, not suitable for entanglement.
There are two ways to insert the insertion point of the electromagnetic flowmeter sensor: one is inserted into the central axis of the pipe to be tested (ie 1/2 times the diameter of the pipe); the other is inserted into the inner wall of the pipe at 1/4 of the pipe.
1. Clean the welding slag and burrs of the base of the pipe to be tested.
2. Turn off the upstream flow control valve or use low pressure water supply.
3. Install the DN50 ball valve on the mounting base. Pay attention to the long cavity of the ball valve. Check whether the ball valve can be fully open and fully closed. Install the screw seat, compression nut and rubber sealing ring onto the ball valve and loosen the positioning nut. Insert the sensor into the rod through the ball valve and insert it into the pipe to be tested. At the same time, it should be noted that the direction of the sensor direction indicator rod should be consistent with the flow direction of the fluid. Ordering Information Please specify the name of the tested medium, working pressure, medium temperature, flow rate (flow rate) range, pipe diameter and the pipe network to be tested.