Vortex flowmeter analysis and solution
3. Reasons for parameter setting direction. The instrument is incorrectly indicated due to a parameter error. The parameter error makes the secondary meter full frequency calculation error, and the reason for this is mainly related to questions 1 and 3. The full-scale frequency is similar, indicating that the long-term inaccuracy is indicated. The full-scale frequency of the actual full-scale frequency and large-dry calculation indicates that the range is fluctuating and cannot be read. The inconsistency of the parameters on the data affects the final determination of the parameters, and finally passes. Recalibration combined with mutual comparison to determine the parameters solves this problem.
4. The secondary instrument is faulty. There are many faults in this part, including: when the instrument board is disconnected, the range setting has individual bit display bad, and the K coefficient setting has individual bit display bad, which makes it impossible to determine the range setting and K factor setting. Part of the reason is mainly related to questions 1, 2. The problem is solved by fixing the corresponding fault.
5, Four-way line connection problem. On the surface of some circuits, the line connection is very good. Check carefully. Some connectors are actually loose and the circuit is interrupted. Some connectors are tightly connected, but the fastening screws are fastened to the wire due to the secondary line problem. Interruption, this part of the reason is mainly related to question.
Gas turbine flowmeter and screwing vortex flowmeter measurement difference
Keywords: gas turbine flowmeter, Tianxin gas turbine flowmeter, gas gas turbine flowmeter
The gas turbine flowmeter has good repeatability and is the preferred flow meter for trade settlement. The gas swirling vortex flowmeter is generally repeatable.
The gas turbine flowmeter can obtain very high frequency signals with strong signal resolution. The frequency of the gas swirling vortex flowmeter is very low, and the smaller the diameter, the lower the signal resolution. The measurement error is to be measured.
The gas turbine flowmeter has a wide range, generally 40:1~20:1, and has a wide measuring range, which can be suitable for occasions with large flow changes. When natural gas is delivered, the flow changes are generally large and, therefore, are particularly suitable for natural gas measurements. The latter range is medium, generally 15:1 to 10:1, and the measurement range is slightly narrower. When measuring gas, it is more suitable for occasions where the flow rate does not change much.
In terms of seismic performance, gas turbine flowmeters have better resistance to pipeline vibration. The gas swirling vortex flowmeter is susceptible to pipeline vibration and should be taken care of when using it.
Battery powered electromagnetic flowmeter features
1. Micro power consumption design, five 3.6V lithium batteries, continuous operation for 5~10 years, and easy to replace the battery.
2. The display adopts LCD large-screen liquid crystal display, which can display multiple flow parameters such as instantaneous flow rate, flow rate (pipe pressure), cumulative total amount of forward and reverse, and alarm prompt.
3. There are no movable and resistance parts in the measuring tube, no entanglement or clogging phenomenon, almost no pressure loss, and long-term reliable and continuous work.
4. Wide measurement range: The flow rate measurement range can reach 0.05m/s~10m/s.
5. High measurement accuracy: ±0.5% ~ ±1.0%.
6. Measurement stability: The measurement accuracy is not affected by changes in physical parameters such as temperature, pressure, viscosity and density of the measured medium.
7, with pulse output and GSM wireless data remote transmission.
8, with data storage function, up to 1000 groups of traffic data can be saved, and the data storage time can be arbitrarily set and data query.