The flowmeter weighing impulse flowmeter that measures the flow using the impulse theorem is used to measure the flow rate of the granular solid medium, and is also used to measure the flow rate of the mud, the crystalline liquid, and the abrasive. Flow measurement ranges from a few kilograms per hour to nearly 10,000 tons. The typical instrument is a horizontal force-separated impulse flowmeter. The measurement principle is that when the measured medium falls freely from a certain height h to a detection plate with a tilt angle, an impulse is generated, and the horizontal force component of the impulse is proportional to the mass flow rate. Measuring this horizontal component can reflect the mass flow.
The electromagnetic flowmeter is made by applying the electric conductor to generate an induced electromotive force in a magnetic field, and the induced electromotive force is proportional to the flow rate, and is measured by measuring the electromotive force to reflect the flow of the pipeline. Its measurement accuracy and sensitivity are high. Industrially used to measure the flow of water, slurry and other media. The maximum diameter can be measured up to 2m, and the pressure loss is extremely small. However, media with low conductivity, such as gas and steam, cannot be used.
The electromagnetic flowmeter has a high cost, and the signal is susceptible to external magnetic field interference, which affects the wide application in industrial tube flow measurement. To this end, the product is constantly improving and updating, to the development of computer.
Liquid turbine flow meter classification and function description
Keywords: liquid turbine flowmeter, diesel liquid turbine flowmeter, pure water liquid turbine flowmeter
First, the liquid turbine flowmeter overview
The liquid turbine flowmeter is the main type of impeller flow (flow rate) meter, and the impeller type flowmeter also has an anemometer and a water meter. The TUF consists of a sensor and a conversion display. The sensor uses a multi-blade rotor to sense the average flow rate of the fluid, thereby deriving the flow or total amount. The rotational speed (or number of revolutions) of the rotor can be detected mechanically, magnetically, and photoelectrically and displayed and transmitted by the reading device.
Second, liquid turbine flowmeter classification
According to the classification of instrument functions, SDLWGY series turbine flowmeters can be divided into 2 categories.
1, turbine flow sensor / transmitter
2, intelligent integrated turbine flowmeter
Third, the liquid turbine flowmeter function description
1, turbine flow sensor / transmitter
This type of turbine flow product itself does not have a live display function, only the flow signal is transmitted far. The flow signal can be divided into pulse signal or current signal (4-20mA); the instrument is low in price, high in integration and small in size, and is especially suitable for use with computer control systems such as secondary display, PLC, DCS.
Plug-in electromagnetic flowmeter features
Corrosion-resistant full stainless steel material, ABS full mold production probe structure is simple, firm, no moving parts, long service life, reliable measurement, strong anti-interference ability, small size, light weight, easy installation, small maintenance range, large measurement range, temperature-free measurement The influence of changes in density, pressure, viscosity, conductivity, etc., zero pressure loss can be installed and disassembled under continuous flow, which brings convenience to the user's maintenance. The manufacturing cost and installation cost are lower than the general electromagnetic flowmeter. It is especially suitable for large and medium-diameter pipeline flow measurement. It adopts advanced processing technology, solid-state packaging, vibration resistance and long life. It makes the meter have good measurement accuracy and stability. The flowmeter not only has 4-20mA current signal output, but also has 1-2000HZ frequency output digital communication output: RS485, RS232, HART, Modbus.