The V-cone flowmeter is a new generation of differential pressure flow measuring instrument. In practical use, many factors have a great influence on the measurement accuracy of the differential pressure flowmeter, which increases the measurement error and reduces the accuracy.
The specific performance is as follows:
1) the design parameters are inconsistent with the working parameters;
2) there is no temperature compensation or compensation is incorrect;
3) the length of the upper and lower straight pipes is insufficient;
4) the cones and pipes are not concentric;
5) the pressure pipes Blockage, etc.
After on-site inspection, the shortage of straight pipe sections, disagreement, and blockage of the pressure guiding pipe can be ruled out.
The differential pressure transmitter is removed for verification, the transmitter is qualified, and the error is within the allowable range.
The technicians recalled the trend record of DCS preservation, and looked at the steam temperature and pressure recording curve. The temperature was between 120 °C and 150 °C, the pressure was between 0.3MPa and 0.6MPa, and the measured temperature and pressure deviated from the design value. The design parameter was temperature. 193 ° C, pressure 1.13 MPa). At the same time, the DCS configuration was viewed and it was found that there was no temperature and pressure compensation for the steam flow in the configuration.
Third, gas turbine flowmeter daily inspection
1. Look at the display: Observe the total amount, instantaneous, temperature, pressure, and battery power. If the abnormality is reported in time.
2. Listen to the running sound of the flowmeter. (Method) The flowmeter should have no noise when it is running normally. If there is a large abnormal noise, it can be judged as damage to the movement of the movement (for example). It needs to be reported and processed in time.
3, data comparison, the output of the first 3 to 5 days before the statistics, if there is no major change, the flowmeter runs normally, such as the successive decrement phenomenon, the gas turbine flowmeter impeller (shaft, shaft core), filtration There is a problem with the device. Need to be reported in time for processing
4, gas turbine flowmeter filling lubricant
(1) Instrument refueling is an important part of the daily maintenance work of the instrument. Lubricating oil is mainly for instrument bearing; lubricating oil can protect the instrument bearing, reduce friction, reduce wear and prolong the operation cycle of the instrument;
(2) Regularly adding lubricating oil can effectively reduce the maintenance rate of the instrument and reduce the maintenance cost;
(3) Ensure the measurement accuracy of the flowmeter. "Measuring is counting money." why would you said this? The management of production and operation accounting, energy saving, cost accounting, etc. is an important production link of enterprise management. If the measurement is not in place, it will affect the economic efficiency and reputation of the enterprise. Therefore, the daily maintenance of the instrument is very important.
Battery powered electromagnetic flowmeter features
1. Micro power consumption design, five 3.6V lithium batteries, continuous operation for 5~10 years, and easy to replace the battery.
2. The display adopts LCD large-screen liquid crystal display, which can display multiple flow parameters such as instantaneous flow rate, flow rate (pipe pressure), cumulative total amount of forward and reverse, and alarm prompt.
3. There are no movable and resistance parts in the measuring tube, no entanglement or clogging phenomenon, almost no pressure loss, and long-term reliable and continuous work.
4. Wide measurement range: The flow rate measurement range can reach 0.05m/s~10m/s.
5. High measurement accuracy: ±0.5% ~ ±1.0%.
6. Measurement stability: The measurement accuracy is not affected by changes in physical parameters such as temperature, pressure, viscosity and density of the measured medium.
7, with pulse output and GSM wireless data remote transmission.
8, with data storage function, up to 1000 groups of traffic data can be saved, and the data storage time can be arbitrarily set and data query.