ltrasonic flowmeter installation
The time difference ultrasonic flowmeter is a competitive flow measurement method in the world today, and its measurement line accuracy is higher than 1.0%. Due to the diversity of the industrial site, especially the environment around the pipeline, how to install and debug the ultrasonic flowmeter according to the specific environment has become an important topic in the field of ultrasonic flow measurement. This procedure explains the installation details of the ultrasonic flowmeter; It further fully demonstrates the advantages of accuracy, reliability and stability of the ultrasonic flowmeter, greatly reducing future maintenance work and even maintenance-free.
Ultrasonic flowmeters should be aware of the site prior to installation, including:
1. What is the distance from the host at the installation of the sensor;
2. Pipe material, pipe wall thickness and pipe diameter;
3. Year of the pipeline;
4. Type of fluid, whether it contains impurities, bubbles, and whether it is full;
5. Fluid temperature
6. Whether there is interference source at the installation site (such as frequency conversion, strong magnetic field, etc.);
7. The temperature of the four seasons at the host site;
8. Whether the power supply voltage used is stable;
9. Whether remote signals and types are required;
According to the site conditions provided above, the manufacturer can configure the site conditions and, if necessary, special models.
Intelligent GPRS MAG flow meter working principle
Since the GPRS communication is a data packet communication network based on an IP address, after the monitoring center computer logs in to the Internet network, the IP address is assigned by the Internet, and each GPRS monitoring point establishes a connection with the host through the address and communicates. Each monitoring point collects data through the data acquisition module, processes the data through the embedded processor in the GPRS terminal, encapsulates the protocol, sends it to the GPRS network, and transmits it to the user data monitoring center computer.
Battery-powered converter GPRS communication introduction
The battery-powered converter with GPRS function can directly send data to the remote control center through the GPRS wireless module GPRS-DY or GPRS-DF, hereinafter referred to as the GPRS module, and then store the collected data in the database at regular intervals. Through the system, even in an off-site away from the observation site, it is convenient to collect and read data of each monitoring point, and truly realize the functions of remote monitoring and data sharing.
In the case of small flow metering, the film surface ratio is the largest, the Roots flowmeter is second, and the turbine flowmeter is the smallest. The initial flow performance of the volumetric flowmeter should be superior to the velocity flowmeter. For industrial and commercial users with large total gas consumption and low heat load for each gas, the volumetric flowmeter, ie, the membrane meter and the Roots flowmeter, should be preferred to meet the flowmeter requirements of a single burner when used alone. It can also meet the measurement requirements of the maximum flow rate when all burners are used at the same time.
Not all industrial and commercial users can solve the measurement problem by selecting a volumetric flowmeter with a larger range. The volumetric ratio of the volumetric flowmeter is limited, and it is impossible to increase without limit. The selection of the flowmeter cannot be solved. This kind of user has a measurement problem. When the user only uses a single device for gas, or when operating at low flow rate, the initial flow rate and minimum flow rate of a single flow meter may not meet the minimum flow rate requirement. A single large flow meter cannot measure low gas consumption, which will result in Large supply and marketing differences.