The installation position and installation method of the ultrasonic flowmeter.
Selecting the installation pipe segment has a great influence on the test accuracy. The selected pipe segment should avoid the interference and eddy current, which have great influence on the measurement accuracy. Generally, the pipe segment should meet the following conditions:
1. Avoid installing the machine in the pump, high-power radio, frequency conversion, that is, where there is strong magnetic field and vibration interference;
2. Select the pipe section where the pipe should be uniform and dense, and it is easy to transmit ultrasonic waves;
3, to have a long straight pipe section, the upstream straight pipe section of the installation point must be greater than 10D (Note: D = diameter), downstream is greater than 5D;
4, the installation point upstream distance pump should have a distance of 30D;
5. The fluid should be filled with pipes;
6. There should be enough space around the pipeline to facilitate the operation of the on-site personnel. The underground pipeline needs to be a test well. The test well is as follows:
Ultrasonic flowmeters generally have two types of probe installation methods, namely Z method and V method.
However, when D < 200mm and the site condition is one of the following conditions, it can also be installed by the Z method:
1. When the measured fluid has high turbidity, when the V method is used to measure the signal or the signal is weak;
2. When the inner wall of the pipe is lined;
3. When the service life of the pipeline is too long and the inner wall is fouled seriously;
For those with better pipeline conditions, even if D is slightly larger than 200mm, in order to improve the measurement accuracy, the V method can be used for installation.
Intelligent GPRS MAG flow meter
The GPRS wireless transmission flowmeter is specially designed for the industry. It can meet the monitoring and management of traffic in urban water supply, heating and gas supply systems, and the processing and calculation of data. It is widely used in urban water supply, heating, gas supply systems, sewage treatment, Water conservancy projects and other industries.
GPRS wireless transmission flowmeter battery-powered converter is to use GPRS wireless communication to achieve remote monitoring. The data of the battery-powered converter can be read in any area covered by the GPRS wireless network. Through the GPRS wireless communication flowmeter network system, the data collected by the field instrument can be transmitted to the monitoring center in real time to achieve unified monitoring and distributed management of the field instrument.
GPRS wireless transmission flowmeter adopts specially designed sensor excitation system and high-performance lithium battery power supply system, and adopts 16-bit embedded ultra-micro power processor, full digital signal processing, stable measurement, high measurement accuracy and strong anti-interference ability. Features such as automatic bidirectional flow measurement, on-site instantaneous flow, positive and negativecumulative total display, self-diagnosis fault alarm, GSM data infinity transfer, etc. Users can realize traffic data through powerful GSM/GPRS remote management software system. GPRS wireless communication, storage, data analysis and other functions.
Plug-in electromagnetic flowmeter features
Corrosion-resistant full stainless steel material, ABS full mold production probe structure is simple, firm, no moving parts, long service life, reliable measurement, strong anti-interference ability, small size, light weight, easy installation, small maintenance range, large measurement range, temperature-free measurement The influence of changes in density, pressure, viscosity, conductivity, etc., zero pressure loss can be installed and disassembled under continuous flow, which brings convenience to the user's maintenance. The manufacturing cost and installation cost are lower than the general electromagnetic flowmeter. It is especially suitable for large and medium-diameter pipeline flow measurement. It adopts advanced processing technology, solid-state packaging, vibration resistance and long life. It makes the meter have good measurement accuracy and stability. The flowmeter not only has 4-20mA current signal output, but also has 1-2000HZ frequency output digital communication output: RS485, RS232, HART, Modbus.