Some experiences on using ultrasonic flowmeter
1. The flowmeter has two modes, diagonal and reflective. When the reflection mode is not detected, it can be measured in diagonal mode, so that we have been using the reflection mode.
2. The flowmeter requires a relatively high pipe, and the insulation layer must be scraped off, otherwise it cannot be measured. When we measured the air conditioning water system, we cut the insulation layer with a knife to prevent the sensor. After the measurement, the cut insulation layer was glued. The surface of the pipe should be as bright as possible. If it is too rough, it must be sanded.
3. It is difficult to measure when the fluid in the pipeline is not full, so the measuring position should be as straight as possible, away from elbows, valves and other places.
4. The authenticity of the flowmeter reading depends on the signal strength. When the signal is too low, the result is basically unreliable, generally more than 60% or even more.
5. Since the readings may vary greatly, the approach we take is to have the flowmeter read continuously, such as continuously recording one minute of reading and then taking the average.
6. Measuring hot water lines is more difficult than cold water lines. Because the wall temperature of the hot water pipe is high, the coupling agent is easily formed at a high temperature. In addition to the product's own coupling agent, we have tried toothpaste.
The summary of ultrasonic flowmeter measurement accuracy and accuracy needs to be improved.
Intelligent GPRS MAG flow meter features
☆ Timing acquisition of electromagnetic flowmeter data: The electromagnetic flowmeter is used to send a data mode to the host every hour.
The data includes:
(1) Instrument status part: module address, module name, current module time, module battery status;
(2) Meter data part: instantaneous flow rate, instantaneous flow rate, fluid conductivity ratio, positive and negative cumulative value
☆ Support short message, fixed IP and dynamic IP address transmission. Data transmission is stable and reliable, low power consumption, and low battery alarm. SMS (SMS): Support all China Unicom and China Mobile SIM
card users; GPRS mode: support for the opening of GPRS wireless Internet access (GPRS access point must be CMNET) China Mobile SIM card users
☆ GPRS communication supports two modes of TCP and UDP, and supports two connection modes of IP address and dynamic domain name.
☆ Integration of acquisition and transmission, improving system reliability and reducing costs;
☆ Do not crash, automatically restore after dropped calls;
☆ GPRS remote online setting and short message remote setting function;
☆ Onboard industrial grade GSM/GPRS communication module, convenient for users to choose GSM/GPRS networking mode
☆ Provide user setting software, open interface
☆ Good waterproof performance
Turbine flow meter applications
Turbine flow meters are widely used in the following measurement objects: petroleum, organic liquids, inorganic liquids, liquefied gases, natural gas, gas and cryogenic fluids. In the transshipment and gathering stations of foreign liquefied petroleum gas, refined oil and light crude oil, the first and last stations of large crude oil transmission pipelines use it for trade settlement. In Europe and the United States, the turbine flowmeter is the natural gas meter next to the orifice flowmeter. In the Netherlands alone, more than 2,600 gas turbine flowmeters of various sizes and pressures from 0.8 MPa to 6.5 MPa are used on the natural gas pipeline. Has become an excellent natural gas flow meter. Although the excellent metering characteristics of the turbine flow meter are favored by people, the impression is that the moving parts are short in use time, and they are inevitable in the selection. After people's unremitting efforts, it should be said that the situation has changed a lot. As the most versatile flowmeter, the turbine flowmeter has been developed into a multi-variety, full-series, multi-size mass production scale.